| ||Format||Pages||Price|| |
|20||$54.00||  ADD TO CART|
|Hardcopy (shipping and handling)||20||$54.00||  ADD TO CART|
|Standard + Redline PDF Bundle||40||$65.00||  ADD TO CART|
Significance and Use
5.1 The total evaporation method is used to measure the isotopic composition of uranium and plutonium materials, and may be used to measure the elemental concentrations of the two elements when employing the IDMS technique.
5.2 Uranium and plutonium compounds are used as nuclear reactor fuels. In order to be suitable for use as a nuclear fuel the starting material must meet certain criteria, such as found in Specifications , , , , , , , , or as specified by the purchaser. The uranium concentration, plutonium concentration, or both, and isotope abundances are measured by thermal ionization mass spectrometry following this method.
5.3 The total evaporation method allows for a wide range of sample loading with no significant change in precision or accuracy. The method is also suitable for trace-level loadings with some loss of precision and accuracy. The total evaporation method and modern instrumentation allow for the measurement of minor isotopes using ion counting detectors, while the major isotope(s) is(are) simultaneously measured using Faraday cup detectors.
5.4 The new generation of miniaturized ion counters allow extremely small samples, in the picogram range, to be measured via the total evaporation method. The method may be employed for measuring environmental or safeguards inspection samples containing nanogram quantities of uranium or plutonium. Very small loadings require special sample handling and careful evaluation of measurement uncertainties.
5.5 Typical uranium analyses are conducted using sample loadings between 50 nanograms and several micrograms. For uranium isotope ratios the total evaporation method had been used in several recent NBL isotopic certified reference material (CRM) characterizations (for example (). A detailed comparison of the total evaporation data on NBL uranium CRMs analyzed by the MAT 261 and TRITONTM instruments is provided in Ref , )(. For total evaporation, plutonium analyses are generally conducted using sample loads in the range of 30 to 400 nanograms of plutonium. )
1.1 This method describes the determination of the isotopic composition, or the concentration, or both, of uranium and plutonium as nitrate solutions by the total evaporation method using a thermal ionization mass spectrometer (TIMS) instrument. Purified uranium or plutonium nitrate solutions are deposited onto a metal filament and placed in the mass spectrometer. Under computer control, ion currents are generated by heating of the filament(s). The ion currents are continually measured until the whole sample is exhausted. The measured ion currents are integrated over the course of the measurement and normalized to a reference isotope ion current to yield isotope ratios.
1.2 In principle, the total evaporation method should yield isotope ratios that do not require mass bias correction. In practice, samples may require this bias correction. Compared to the conventional TIMS method described in Test Method , the total evaporation method is approximately two times faster, improves precision of the isotope ratio measurements by a factor of two to four, and utilizes smaller sample sizes. Compared to the method, the total evaporation method provides “major” isotope ratios 235U/238U and 240Pu/239Pu with improved accuracy.
1.3 The total evaporation method is prone to biases in the “minor” isotope ratios (233U/238U, 234U/238U, and 236U/238U ratios for uranium materials and 238Pu/239Pu, 241Pu/239Pu, 242Pu/239Pu, and 244Pu/239Pu ratios for plutonium materials) due to peak tailing from adjacent major isotopes. The magnitude of the absolute bias is dependent on measurement and instrumental characteristics. The relative bias, however, depends on the relative isotopic abundances of the sample. The use of an electron multiplier equipped with an energy filter may eliminate or diminish peak tailing effects. Measurement of the abundance sensitivity of the instrument may be used to ensure that such biases are negligible, or may be used to bias correct the minor isotope ratios.
1.4 The values stated in SI units are to be regarded as standard. When non-SI units are provided in parentheses, they are for information only.
1.5 This standard may involve the use of hazardous materials and equipment. This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C753 Specification for Nuclear-Grade, Sinterable Uranium Dioxide Powder
C757 Specification for Nuclear-Grade Plutonium Dioxide Powder for Light Water Reactors
C776 Specification for Sintered Uranium Dioxide Pellets
C787 Specification for Uranium Hexafluoride for Enrichment
C833 Specification for Sintered (Uranium-Plutonium) Dioxide Pellets for Light Water Reactors
C859 Terminology Relating to Nuclear Materials
C967 Specification for Uranium Ore Concentrate
C996 Specification for Uranium Hexafluoride Enriched to Less Than 5% 235U
C1008 Specification for Sintered (Uranium-Plutonium) DioxidePellets--Fast Reactor Fuel
C1068 Guide for Qualification of Measurement Methods by a Laboratory Within the Nuclear Industry
C1156 Guide for Establishing Calibration for a Measurement Method Used to Analyze Nuclear Fuel Cycle Materials
C1168 Practice for Preparation and Dissolution of Plutonium Materials for Analysis
C1347 Practice for Preparation and Dissolution of Uranium Materials for Analysis
C1411 Practice for The Ion Exchange Separation of Uranium and Plutonium Prior to Isotopic Analysis
C1415 Test Method for 238Pu Isotopic Abundance By Alpha Spectrometry
C1625 Test Method for Uranium and Plutonium Concentrations and Isotopic Abundances by Thermal Ionization Mass Spectrometry
C1816 Practice for The Ion Exchange Separation of Small Volume Samples Containing Uranium, Americium, and Plutonium Prior to Isotopic Abundance and Content Analysis
C1832 Test Method for Determination of Uranium Isotopic Composition by the Modified Total Evaporation (MTE) Method Using a Thermal Ionization Mass Spectrometer
D1193 Specification for Reagent Water
D3084 Practice for Alpha-Particle Spectrometry of Water
ICS Number Code 27.120.30 (Fissile materials and nuclear fuel technology)
UNSPSC Code 15131500(Nuclear fuel)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM C1672-17, Standard Test Method for Determination of Uranium or Plutonium Isotopic Composition or Concentration by the Total Evaporation Method Using a Thermal Ionization Mass Spectrometer, ASTM International, West Conshohocken, PA, 2017, www.astm.orgBack to Top