| ||Format||Pages||Price|| |
|5||$55.20||  ADD TO CART|
This test method covered the determination of the concentration of uranium-235 and uranium-238 in urine using Inductively Coupled Plasma-Mass Spectrometry. This test method could have been used to support uranium facility bioassay programs.
Formerly under the jurisdiction of Committee C26 on Nuclear Fuel Cycle, this test method was withdrawn in June 2017 and replaced by Test Methodon the Determination of Uranium in Urine by Inductively Coupled Plasma Mass Spectrometer Following Nitric Acid Dilution.
1.1 This test method covers the determination of the concentration of uranium-235 and uranium-238 in urine using Inductively Coupled Plasma-Mass Spectrometry. This test method can be used to support uranium facility bioassay programs.
1.2 This method detection limits for 235U and 238U are 6 ng/L. To meet the requirements of ANSI N13.30, the minimum detectable activity (MDA) of each radionuclide measured must be at least 0.1 pCi/L (0.0037 Bq/L). The MDA translates to 47 ng/L for 235U and 300 ng/L for 238U. Uranium– 234 cannot be determined at the MDA with this test method because of its low mass concentration level equivalent to 0.1 pCi/L.
1.3 The digestion and anion separation of urine may not be necessary when uranium concentrations of more than 100 ng/L are present.
1.4 Units—The values stated in picoCurie per liter units are to be regarded as standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Note 1—Warning: The ICP-MS is a source of intense ultraviolet radiation from the radio frequency induced plasma. Protection from radio frequency radiation and UV radiation is provided by the instrument under normal operation.
Note 1—Warning: The ICP-MS is a source of intense ultraviolet radiation from the radio frequency induced plasma. Protection from radio frequency radiation and UV radiation is provided by the instrument under normal operation.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C859 Terminology Relating to Nuclear Materials
C1310 Test Method for Determining Radionuclides in Soils by Inductively Coupled Plasma-Mass Spectrometry Using Flow Injection Preconcentration
C1345 Test Method for Analysis of Total and Isotopic Uranium and Total Thorium in Soils by Inductively Coupled Plasma-Mass Spectrometry
D1193 Specification for Reagent Water
Other DocumentsDOE Order 5480.11 Radiological Measurements Quality
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM C1379-10, Standard Test Method for Analysis of Urine for Uranium-235 and Uranium-238 Isotopes by Inductively Coupled Plasma-Mass Spectrometry (Withdrawn 2017), ASTM International, West Conshohocken, PA, 2010, www.astm.orgBack to Top