| ||Format||Pages||Price|| |
|11||$54.00||  ADD TO CART|
|Hardcopy (shipping and handling)||11||$54.00||  ADD TO CART|
Significance and Use
5.1 This test method is useful for determining the plutonium content of scrap and waste in containers ranging from small cans with volumes of the order of a mL to crates and boxes of several thousand liters in volume. A common application would be to 208-L (55-gal) drums. Total Pu content ranges from 10 mg to 6 kg (. The upper limit may be restricted depending on specific matrix, calibration material, criticality safety, or counting equipment considerations. )
5.2 This test method is applicable for U.S. Department of Energy shipper/receiver confirmatory measurements (, nuclear material diversion detection, and International Atomic Energy Agency attributes measurements )(. )
5.3 This test method should be used in conjunction with a scrap and waste management plan that segregates scrap and waste assay items into material categories according to some or all of the following criteria: bulk density, the chemical forms of the plutonium and the matrix, americium to plutonium isotopic ratio, and hydrogen content. Packaging for each category should be uniform with respect to size, shape, and composition of the container. Each material category might require calibration standards and may have different Pu mass limits.
5.4 Bias in passive neutron coincidence measurements is related to item size and density, the homogeneity and composition of the matrix, and the quantity and distribution of the nuclear material. The precision of the measurement results is related to the quantity of nuclear material, the (α,n) reaction rate, and the count time of the measurement.
5.4.1 For both benign matrix and matrix specific measurements, the method assumes the calibration reference materials match the items to be measured with respect to the homogeneity and composition of the matrix, the neutron moderator and absorber content, and the quantity of nuclear material, to the extent they affect the measurement.
5.4.2 Measurements of smaller containers containing scrap and waste are generally more accurate than measurements of larger items.
5.4.3 It is recommended that where feasible measurements be made on items with homogeneous contents. Heterogeneity in the distribution of nuclear material, neutron moderators, and neutron absorbers have the potential to cause biased results.
5.5 The coincident neutron production rates measured by this test method are related to the mass of the even number isotopes of plutonium. If the relative abundances of these isotopes are not accurately known, biases in the total Pu assay value will result.
5.6 Typical count times are in the range of 300 to 3600 s.
5.7 Reliable results from the application of this method require training of the personnel who package the scrap and waste prior to measurement and of personnel who perform the measurements. Training guidance is available from ANSI 15.20, Guides , , , and .
1.1 This test method describes the nondestructive assay of scrap or waste for plutonium content using passive thermal-neutron coincidence counting. This test method provides rapid results and can be applied to a variety of carefully sorted materials in containers as large as several thousand liters in volume. The test method applies to measurements of 238Pu, 240Pu, and 242Pu and has been used to assay items whose total plutonium content ranges from 10 mg to 6 kg (. )
1.2 This test method requires knowledge of the relative abundances of the Pu isotopes to determine the total Pu mass (Test Method ).
1.3 This test method may not be applicable to the assay of scrap or waste containing other spontaneously fissioning nuclides.
1.3.1 This test method may give biased results for measurements of containers that include large amounts of hydrogenous materials.
1.3.2 The techniques described in this test method have been applied to materials other than scrap and waste (. , )
1.4 This test method assumes the use of shift-register-based coincidence technology (. )
1.5 Several other techniques that are often encountered in association with passive neutron coincidence counting exist. These include neutron multiplicity counting (, Test Method , ), add-a-source analysis for matrix correction (, flux probes also for matrix compensation, cosmic-ray rejection )( to improve precision close to the detection limit, and alternative data collection electronics such as list mode data acquisition. Passive neutron coincidence counting may also be combined with certain active interrogation schemes as in Test Methods ) and . Discussions of these established techniques are not included in this method.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C986 Guide for Developing Training Programs in the Nuclear Fuel Cycle
C1009 Guide for Establishing and Maintaining a Quality Assurance Program for Analytical Laboratories Within the Nuclear Industry
C1030 Test Method for Determination of Plutonium Isotopic Composition by Gamma-Ray Spectrometry
C1068 Guide for Qualification of Measurement Methods by a Laboratory Within the Nuclear Industry
C1128 Guide for Preparation of Working Reference Materials for Use in Analysis of Nuclear Fuel Cycle Materials
C1133/C1133M Test Method for Nondestructive Assay of Special Nuclear Material in Low-Density Scrap and Waste by Segmented Passive Gamma-Ray Scanning
C1210 Guide for Establishing a Measurement System Quality Control Program for Analytical Chemistry Laboratories Within the Nuclear Industry
C1316 Test Method for Nondestructive Assay of Nuclear Material in Scrap and Waste by Passive-Active Neutron Counting Using252Cf Shuffler
C1458 Test Method for Nondestructive Assay of Plutonium, Tritium and241Am by Calorimetric Assay
C1490 Guide for the Selection, Training and Qualification of Nondestructive Assay (NDA) Personnel
C1493 Test Method for Non-Destructive Assay of Nuclear Material in Waste by Passive and Active Neutron Counting Using a Differential Die-Away System
C1500 Test Method for Nondestructive Assay of Plutonium by Passive Neutron Multiplicity Counting
C1592/C1592M Guide for Making Quality Nondestructive Assay Measurements
C1673 Terminology of C26.10 Nondestructive Assay Methods
ICS Number Code 13.030.30 (Special wastes)
UNSPSC Code 12141735(Plutonium Pu); 76131500(Nuclear waste treatment)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM C1207-10(2018), Standard Test Method for Nondestructive Assay of Plutonium in Scrap and Waste by Passive Neutron Coincidence Counting, ASTM International, West Conshohocken, PA, 2018, www.astm.orgBack to Top