| ||Format||Pages||Price|| |
|5||$55.20||  ADD TO CART|
Formerly under the jurisdiction of Committee B08 on Metallic and Inorganic Coatings, this test method was withdrawn in November 2016. This standard is being withdrawn without replacement due to its limited use by industry.
1.1 This test method covers the measurement of the thickness of transparent metal oxide and metallic coatings by utilizing a double-beam interference microscope.
1.2 The test method requires that the specimen surface or surfaces be sufficiently mirrorlike to form recognizable fringes.
1.3 This test method can be used nondestructively to measure 1 to 10μ m thick transparent coatings, such as anodic coatings on aluminum. The test method is used destructively for 0.1 to 10 μm thick opaque coatings by stripping a portion of the coating and measuring the step height between the coating and the exposed substrate. The stripping method can also be used to measure 0.2 to 10 μm thick anodic coatings on aluminum.
1.4 The test method is usable as a reference method for the measurement of the thickness of the anodic film on aluminum or of metallic coatings when the technique includes complete stripping of a portion of the coating without attack of the substrate. For anodic films on aluminum, the thickness must be greater than 0.4 μm; the uncertainty can be as great as 0.2 μm. For metallic coatings, the thickness must be greater than 0.25 μm; the uncertainty can be as great as 0.1 μm.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
B504 Test Method for Measurement of Thickness of Metallic Coatings by the Coulometric Method
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM B588-88(2010), Standard Test Method for Measurement of Thickness of Transparent or Opaque Coatings by Double-Beam Interference Microscope Technique (Withdrawn 2016), ASTM International, West Conshohocken, PA, 2010, www.astm.orgBack to Top