| ||Format||Pages||Price|| |
|PDF (96K)||5||$25||  ADD TO CART|
|Complete Source PDF (3.6M)||149||$60||  ADD TO CART|
Ten years ago the coefficient of variation of creep test measurements in our laboratory was reported as being 8.6% to 12%. Now it is between 1.0 and 1.6%. This paper reports on the changes which have brought about this large improvement in accuracy.
creep, strain measurement, laboratory practice, repeatability
Technical Executive, ERA Technology Ltd., Leatherhead, Surrey
Technical Officer, ERA Technology Ltd., Leatherhead, Surrey