1.1This test method is intended to quantitatively determine the non-condensed hydrocarbon gases with carbon numbers from C1 to C5+ and non-hydrocarbon gases, such as H2, CO2, O2, N2, and CO, in gaseous samples. This test method is a companion standard test method to ASTM D1945 and ASTM D1946 differing in that it incorporates use of capillary columns instead of packed columns and allows other technological differences. 1.2Hydrogen sulfide can be detected but may not be accurately determined by this procedure due to loss in sample containers or sample lines and possible reactions unless special precautions are taken. 1.3Non-hydrocarbon gases have a lower detection limit in the concentration range of 0.03 to 100 mole percent using a thermal conductivity detector (TCD) detector and C1 to C6 hydrocarbons have a lower detection limit in the range of 0.005 to 100 mole percent using a flame ionization detector (FID); using a TCD may limit quantification of hydrocarbons down to approximately 300 ppm. 1.3.1Hydrocarbon detection limits can be reduced with the use of pre-concentration techniques and/or cryogenic trapping. 1.4This test method does not fully determine individual hydrocarbons heavier than benzene, which are grouped together as C7+ When detailed analysis is not required the compounds with carbon number greater than C5 may be grouped as either C6+, or C7+. Accurate analysis of C5+ components depends on proper vaporization of these compounds during sampling at process unit sources as well as in the sample introduction into the analyzer in the laboratory. 1.5Water vapor may interfere with the C6+ analysis if a TCD detector is used. 1.6Helium may interfere with the hydrogen determination and argon with oxygen. Depending on the analyzer used, pentenes, if present, may either be separated or grouped with the C6+ components. 1.7The values stated in SI units are to be regarded as standard. 1.8This standard does not purport to address all of the safety problems, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulator limitations prior to use.
Keywordsgas chromatography; gas sampling valve; hydrocarbons; nonhydrocarbon gases; gas analysis
This test method is a companion standard test method to ASTM D1945 and ASTM D1946 differing in that it incorporates use of capillary columns instead of packed columns and allows other technological differences.Back to Top
Brice Van Der Volgen