SUPERSEDED (click for Active standard)
| ||Format||Pages||Price|| |
|8||$62.00||  ADD TO CART|
1.1 This test method covers the measurement of the magnetically induced torque produced by the static magnetic field in the magnetic resonance environment on medical devices and the comparison of that torque to the equivalent torque applied by the gravitational force to the implant.
1.2 This test method does not address other possible safety issues which include but are not limited to issues of magnetically induced force due to spatial gradients in the static magnetic field, RF heating, induced heating, acoustic noise, interaction among devices, and the functionality of the device and the MR system.
1.3 The torque considered here is the magneto-static torque due to the interaction of the MRI static magnetic field with the magnetization in the implant. The dynamic torque due to interaction of the static field with eddy currents induced in a rotating device is not addressed in this test method. Currents in lead wires may induce a torque as well.
1.4 The sensitivity of the torque measurement apparatus must be greater than 1/10 the “gravity torque,” the product of the device's maximum linear dimension and its weight.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
F2052 Test Method for Measurement of Magnetically Induced Displacement Force on Medical Devices in the Magnetic Resonance Environment
F2119 Test Method for Evaluation of MR Image Artifacts from Passive Implants
F2182 Test Method for Measurement of Radio Frequency Induced Heating On or Near Passive Implants During Magnetic Resonance Imaging
F2503 Practice for Marking Medical Devices and Other Items for Safety in the Magnetic Resonance Environment
ICS Number Code 17.220.20 (Measurement of electrical and magnetic quantities)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F2213-06(2011), Standard Test Method for Measurement of Magnetically Induced Torque on Medical Devices in the Magnetic Resonance Environment, ASTM International, West Conshohocken, PA, 2011, www.astm.orgBack to Top