Active Standard ASTM G59 | Developed by Subcommittee: G01.11
Book of Standards Volume: 03.02
Historical (view previous versions of standard)
Significance and Use
This test method can be utilized to verify the performance of polarization resistance measurement equipment including reference electrodes, electrochemical cells, potentiostats, scan generators, measuring and recording devices. The test method is also useful for training operators in sample preparation and experimental techniques for polarization resistance measurements.
Polarization resistance can be related to the rate of general corrosion for metals at or near their corrosion potential, Ecorr. Polarization resistance measurements are an accurate and rapid way to measure the general corrosion rate. Real time corrosion monitoring is a common application. The technique can also be used as a way to rank alloys, inhibitors, and so forth in order of resistance to general corrosion.
In this test method, a small potential scan, ΔE(t), defined with respect to the corrosion potential (ΔE = E – Ecorr), is applied to a metal sample. The resultant currents are recorded. The polarization resistance, RP, of a corroding electrode is defined from Eq 1 as the slope of a potential versus current density plot at i = 0 (1-4):
The units of the Tafel slopes are V. The corrosion rate, CR, in mm per year can be determined from Eq 4 in which EW is the equivalent weight of the corroding species in grams and ρ is the density of the corroding material in g/cm3.
Refer to Practice G 102
The test method may not be appropriate to measure polarization resistance on all materials or in all environments. See 9.2 for a discussion of method biases arising from solution resistance and electrode capacitance.
1.1 This test method covers an experimental procedure for polarization resistance measurements which can be used for the calibration of equipment and verification of experimental technique. The test method can provide reproducible corrosion potentials and potentiodynamic polarization resistance measurements.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
G3 Practice for Conventions Applicable to Electrochemical Measurements in Corrosion Testing
G5 Reference Test Method for Making Potentiostatic and Potentiodynamic Anodic Polarization Measurements
G102 Practice for Calculation of Corrosion Rates and Related Information from Electrochemical Measurements
ICS Number Code 17.220.20 (Measurement of electrical and magnetic quantities)
UNSPSC Code 41114604(Corrosion testers)