| ||Format||Pages||Price|| |
|6||$43.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$43.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$51.60||  ADD TO CART|
Significance and Use
This practice applies to and establishes a method to measure the imaging performance of X-ray systems used for the screening for prohibited items such as weapons, explosives and explosive devices in baggage, packages, cargo or mail.
This practice is intended for use by manufacturers to assess performance and by evaluators of security and contraband screening X-ray systems to verify performance.
This practice is intended to establish whether an X-ray system meets the manufacturer's specification or if the system's performance has deteriorated over time.
This practice may be used for manufacturing control, specification acceptance, service evaluation or regulatory statutes.
This practice is intended for use at both the point of manufacture and where the system is operated. The latter includes locations such as security checkpoints of transportation terminals, nuclear power stations, correctional institutions, corporate mailrooms, government offices and other security areas.
The most significant attributes of this practice are the design of a standard Test Object and standard methods for determining the performance levels of the system.
In screening objects with X-ray systems video images are the primary inputs provided to operators. The better the quality of these images, the better the potential performance of the operator.
1.1 This practice applies to all X-ray based screening systems, with tunnel apertures up to 1 m wide × 1 m high, whether it is a conventional X-ray system or an explosives detection system (EDS) that provides a projection or projection/scatter image for an operator to interpret.
1.2 This practice applies to X-ray systems used for the screening for prohibited items such as weapons, explosives, and explosive devices in baggage, packages, cargo, or mail.
1.3 This practice establishes quantitative and qualitative methods for evaluating the systems. This practice does not establish minimum performance requirements for any particular application.
1.4 This practice relies upon the use of a standard test object (ASTM X-ray Test Object) to determine the applicable performance levels of the systems. The specific test object is subsequently described and referred to in this document as the “Test Object.”
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
Other DocumentsHealth&SafetyReg (United Kingdom)
B258 Specification for Nominal Diameters and Cross-Sectional Areas of AWG Sizes of Solid Round Wires Used as Electrical Conductors
E1025 Practice for Design, Manufacture, and Material Grouping Classification of Hole-Type Image Quality Indicators (IQI) Used for Radiology
E1316 Terminology for Nondestructive Examinations
E1647 Practice for Determining Contrast Sensitivity in Radiology
F1039 Test Method for Measurement of Low Level X-Radiation Used in X-Ray Security Screening Systems
ICS Number Code 13.310 (Protection against crime)
UNSPSC Code 42201848(Digital x ray imaging system)
ASTM F792-08, Standard Practice for Evaluating the Imaging Performance of Security X-Ray Systems, ASTM International, West Conshohocken, PA, 2008, www.astm.orgBack to Top