This specification covers polysulfone resin (poly(oxy-p-phenylenesulfonyl-p-phenyleneoxy-p-phenyleneisopropylidene-p-phenylene)) for medical applications. Requirements and associated test methods for a form of this thermoplastic intended for use in manufacturing medical devices or components of medical devices are provided. The use of this resin in medical devices should be restricted to nonimplant applications until biocompatibility evaluations appropriate for the intended applications are successfully completed. The molecular weight of the resin shall be determined by osmotic pressure in monochlorobenzene. The polysulfone resin shall yield an infrared transmittance spectrum that exhibits major transmittance bands only at the same wavelengths as that of a reference spectrum. Medical devices made of polysulfone may be repeatedly sterilized through steam, ethylene oxide, irradiation, and dry heat sterilization, among others. The polysulfone resin shall be tested for nonvolatile content and melt flow, and shall conform to the specified electrical, physical and mechanical, and thermal properties.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers polysulfone resin (poly(oxy-1,4-phenylenesulfonyl-1,4–phenylene (dimethylmethylene)-1,4–phenylene)) as defined in ISO 25137–1, supplied by a vendor in virgin form (pellets, powder, fabricated forms and so forth) for medical applications. This specification provides requirements and associated test methods for this thermoplastic when it is intended for use in manufacturing medical devices or components of medical devices.
1.2 As with any material, some characteristics may be altered by the processing techniques (such as molding, extrusion, machining, sterilization, and so forth) required for the production of a specific part or device. Therefore, properties of fabricated forms of this resin should be evaluated using test methods which are appropriate to ensure safety and efficacy as agreed upon by the vendor, purchaser, and regulating bodies.
1.3 The standard allows for designation of polysulfone resin for all medical applications. The actual extent of performance and suitability for a specific application must be evaluated by the vendor, purchaser, and regulating bodies.
1.4 The properties included in this specification are those applicable for unfilled polysulfone (PSU) polymers with the addition of colorants and processing aids. Indicated properties are for injection molded forms. Forms containing fillers or other additives, as well as polymer blends which contain PSU, or reclaimed materials, are not covered by this specification.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 When evaluating material in accordance with this specification, hazardous materials, operations, and equipment may be involved. This standard does not purport to address all of the concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D256 Test Methods for Determining the Izod Pendulum Impact Resistance of Plastics
D638 Test Method for Tensile Properties of Plastics
D648 Test Method for Deflection Temperature of Plastics Under Flexural Load in the Edgewise Position
D792 Test Methods for Density and Specific Gravity (Relative Density) of Plastics by Displacement
D6394 Specification for Sulfone Plastics (SP)
F748 Practice for Selecting Generic Biological Test Methods for Materials and Devices
plastic surgical devices/applications; polymerssurgical applications; polysulfone resins ; unfilled polysulfone (PSU); Plastic surgical devices/applications--specifications; Polymers (surgical applications)--specifications; Polysulfone resins; Surgical implant resins--specifications; Surgical implants--specifications; Thermoplastic surgical implant applications--specifications ;
ICS Number Code 11.120.01 (Pharmaceutics in general); 83.080.20 (Thermoplastic materials)
ASTM International is a member of CrossRef.
Citing ASTM Standards
[Back to Top]