1.1 This specification covers requirements of wrought 35cobalt-35nickel-20chromium-10molybdenum alloy (UNS R30035) in the form of bar and wire, used for the manufacture of surgical implants. This alloy depends on combinations of work strengthening and aging to attain a variety of combinations of strength and ductility.
1.2 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
Society of Automotive Engineers
SAE J1086 Practice for Numbering Metals and Alloys (UNS)
American Society for Quality Standard
ASQ C1 Specification of General Requirements for a Quality Program Available from American Society for Quality, 600 N. Plankinton Ave., Milwaukee, WI 53203.
A751 Test Methods, Practices, and Terminology for Chemical Analysis of Steel Products
E8 Test Methods for Tension Testing of Metallic Materials
E29 Practice for Using Significant Digits in Test Data to Determine Conformance with Specifications
E112 Test Methods for Determining Average Grain Size
E354 Test Methods for Chemical Analysis of High-Temperature, Electrical, Magnetic, and Other Similar Iron, Nickel, and Cobalt Alloys
F981 Practice for Assessment of Compatibility of Biomaterials for Surgical Implants with Respect to Effect of Materials on Muscle and Bone
ISO 9001 Quality Management Systems
cobalt alloys (for surgical implants) cobalt-nickel-chromium-molybdenum alloy; metals (for surgical implants)-cobalt alloys
ASTM International is a member of CrossRef.
Citing ASTM Standards
[Back to Top]