| ||Format||Pages||Price|| |
|14||$51.00||  ADD TO CART|
|Hardcopy (shipping and handling)||14||$51.00||  ADD TO CART|
Significance and Use
4.1 The purpose of this guide is to provide guidance for selecting appropriate device size(s) and determining appropriate sample size(s) for design verification of endovascular devices. The device size(s) and sample size(s) for each design input requirement should be determined before testing. The device size(s) selected for verification testing should establish that the entire device matrix is able to achieve the design input requirements. If testing is not performed on all device sizes, justification should be provided.
4.2 The sample size justification and statistical procedures used to analyze the data should be based on sound scientific principles and should be suitable for reaching a justifiable conclusion. Insufficient sample size may lead to erroneous conclusions more often than desired.
4.3 Guidance regarding methodologies for determining device size selection and appropriate sample size is provided in Sections and .
1.1 This guide provides guidance for selecting an appropriate device size(s) and determining an appropriate sample size(s) (that is, number of samples) for design verification testing of endovascular devices. A methodology is presented to determine which device size(s) should be selected for testing to verify the device design adequately for each design input requirement (that is, test characteristic). Additionally, different statistical approaches are presented and discussed to help guide the developer to determine and justify sample size(s) for the design input requirement being verified. Alternate methodologies for determining device size selection and sample size selection may be acceptable for design verification.
1.2 This guide applies to physical design verification testing. This guide addresses in-vitro testing; in-vivo/animal studies are outside the scope of this guide. This guide does not directly address design validation; however, the methodologies presented may be applicable to in-vitro design validation testing. Guidance for sampling related to computational simulation (for example, sensitivity analysis and tolerance analysis) is not provided. Guidance for using models, such as design of experiments (DOE), for design verification testing is not provided. This guide does not address sampling across multiple manufacturing lots as this is typically done as process validation. Special considerations are to be given to certain tests such as fatigue (see Practice ) and shelf life testing (see Section ).
1.3 Regulatory guidance may exist for endovascular devices that should be considered for design verification device size and sample size selection.
1.4 Units—The values stated in SI units are to be regarded as the standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E739 Practice for Statistical Analysis of Linear or Linearized Stress-Life (S-N) and Strain-Life (-N) Fatigue Data
F2914 Guide for Identification of Shelf-life Test Attributes for Endovascular Devices
ISO StandardsISO 14971:2012
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F3172-15, Standard Guide for Design Verification Device Size and Sample Size Selection for Endovascular Devices, ASTM International, West Conshohocken, PA, 2015, www.astm.orgBack to Top