Active Standard ASTM F2886 | Developed by Subcommittee: F04.12
Book of Standards Volume: 13.02
This specification covers chemical, mechanical, and metallurgical general requirements for metal injection molded (MIM) cobalt-28chromium-6molybddenum components to be used in manufacturing surgical implants. In this specification, the MIM components covered may have been densified beyond their as-sintered density by post-sinter processing. For the chemical requirements, the components supplied in this specification must conform in accordance to the chemical requirements specified herein in Table 1. The product analysis tolerances must also conform to the product tolerances presented in Table 2. The specification also enumerates the mechanical requirements for MIM components wherein the tensile properties of the MIM must conform to the mechanical properties in Table 3. The microstructural requirements and specimen preparation shall be in accordance with Guide E3 and Practice E407.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers the chemical, mechanical, and metallurgical requirements for metal injection molded (MIM) cobalt-28chromium-6molybdenum components to be used in the manufacture of surgical implants
1.2 The MIM components covered by this specification may have been densified beyond their as-sintered density by post-sinter processing.
1.3 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
B243 Terminology of Powder Metallurgy
B311 Test Method for Density of Powder Metallurgy (PM) Materials Containing Less Than Two Percent Porosity
B923 Test Method for Metal Powder Skeletal Density by Helium or Nitrogen Pycnometry
E3 Guide for Preparation of Metallographic Specimens
E8/E8M Test Methods for Tension Testing of Metallic Materials
E29 Practice for Using Significant Digits in Test Data to Determine Conformance with Specifications
E165 Practice for Liquid Penetrant Examination for General Industry
E354 Test Methods for Chemical Analysis of High-Temperature, Electrical, Magnetic, and Other Similar Iron, Nickel, and Cobalt Alloys
E407 Practice for Microetching Metals and Alloys
F75 Specification for Cobalt-28 Chromium-6 Molybdenum Alloy Castings and Casting Alloy for Surgical Implants (UNS R30075)
F601 Practice for Fluorescent Penetrant Inspection of Metallic Surgical Implants
F629 Practice for Radiography of Cast Metallic Surgical Implants
F1537 Specification for Wrought Cobalt-28Chromium-6Molybdenum Alloys for Surgical Implants (UNS R31537, UNS R31538, and UNS R31539)
SI10 American National Standard for Use of the International System of Units (SI): The Modern Metric System
ICS Number Code 11.040.40 (Implants for surgery, prothetics and orthotics)
UNSPSC Code 42295500(Surgical implants and expanders and extenders and surgical wires and related products)