| ||Format||Pages||Price|| |
|23||$62.00||  ADD TO CART|
|Hardcopy (shipping and handling)||23||$62.00||  ADD TO CART|
Significance and Use
This test method is intended to assess a combination of inherent chemical irritation and mechanical irritation for products and materials expected to come into contact with the skin. It is a comparative approach whereby the potential irritation of a test material is compared to that of a reference material similar in form and composition. The reference material should have a known safety and irritation profile.
1.1 The Behind-the-Knee (BTK) method, using the popliteal fossa of human volunteers as a test site, simultaneously evaluates the inherent chemical irritation, and the potential for mechanical irritation of substrates and products that are designed to come into repeated or extended close contact with the skin (see validation references 1-7). This is a bilateral test comparing a test material to a reference material with a known safety profile.
1.2 This test method shall be used by qualified health care professionals experienced in good clinical practice (GCP) procedures.
1.3 This test method can be performed using human subjects on either intact or compromised skin. Testing should be performed on intact skin for test substrates or products expected to have contact with normal, intact skin, or for direct comparison to products with a known skin irritation profile. Testing can be performed on compromised skin for test substrates or products that may commonly come into contact with damaged skin (for example, skin with diaper rash, or chapped skin) or skin that is expected to be hydrated.
1.4 Visual scoring of erythema and dryness is performed by a trained skin grader on a pre-defined scale.
1.5 Prior to use in this test, materials shall undergo overall favorable biocompatibility testing consistent with the approach outlined in protocol Practice F748 or ISO 10993-1:2009. As a part of this series of testing, irritation per Practice F719 or ISO 10993-10 shall be conducted.
1.6 The values stated in inch-pound units are to be regarded as standard. No other units of measurement are included in this standard.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D6355 Test Method for Human Repeat Insult Patch Testing of Medical Gloves
F719 Practice for Testing Biomaterials in Rabbits for Primary Skin Irritation
F748 Practice for Selecting Generic Biological Test Methods for Materials and Devices
ISO StandardsISO 10993-10 Biological Evaluation of Medical Devices--Part 10: Tests for Irritation and Delayed-type Hypersensitivity
ICS Number Code 11.100.99 (Other standards related to laboratory medicine)
UNSPSC Code 42321800(Knee joint implants)
ASTM F2808-10, Standard Test Method for Performing Behind-the-Knee (BTK) Test for Evaluating Skin Irritation Response to Products and Materials That Come Into Repeated or Extended Contact with Skin, ASTM International, West Conshohocken, PA, 2010, www.astm.orgBack to Top