| ||Format||Pages||Price|| |
|5||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$44.00||  ADD TO CART|
Significance and Use
5.1 The O2GTR is an important determinant of the packaging protection afforded by barrier materials. It is not, however, the sole determinant, and additional tests, based on experience, must be used to correlate packaging performance with O2GTR. It is suitable as a referee method of testing, provided that the purchaser and the seller have agreed on sampling procedures, standardization procedures, test conditions, and acceptance criteria.
5.2 Testing which has compared select instruments with other sensors to the instruments specifically described in Test Method D3985 is shown in Section 16, Precision and Bias, of this method.
5.3 The Precision and Bias section of this method shows results using several instruments with non-coulometric and coulometric sensors.
1.1 This test method covers a procedure for determination of the steady-state rate of transmission of oxygen gas through plastics in the form of film, sheeting, laminates, coextrusions, or plastic-coated papers or fabrics. It provides for the determination of (1) oxygen gas transmission rate (O2GTR), (2) the permeance of the film to oxygen gas (PO2), and (3) oxygen permeability coefficient (P'O2) in the case of homogeneous materials.
1.2 This test method does not purport to be the only method for measurement of O2GTR. There may be other methods of O2GTR determination that use other oxygen sensors and procedures.
1.3 This test method has intentionally been prepared to allow for the use of various sensors, devices, and procedures. The precision and bias of each design needs to be individually established to determine the applicability of that instrument or method to meet the needs of the user.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1898 Practice for Sampling of Plastics
D3985 Test Method for Oxygen Gas Transmission Rate Through Plastic Film and Sheeting Using a Coulometric Sensor
ICS Number Code 83.140.10 (Films and sheets)
UNSPSC Code 30265802(Plastic sheet)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F2622-08(2013), Standard Test Method for Oxygen Gas Transmission Rate Through Plastic Film and Sheeting Using Various Sensors, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top