Active Standard ASTM F2559 / F2559M | Developed by Subcommittee: F02.50
Book of Standards Volume: 15.10
Historical (view previous versions of standard)
Significance and Use
Medical device peel pouches are universally used by the industry and produced by a myriad of suppliers. They may be constructed of many different materials including films, foils, paper, nonwovens such as Tyvek, and combinations thereof. However, even with the diversity of materials, there are still basic requirements that all pouches should exhibit. Above all, the pouches must contain and protect the device while maintaining sterility during all physical handling.
Pouch requirements may be divided into two categories, initial pouch and material qualification, and routine production and receipt requirements to ensure the purchaser receives exactly what is ordered. While all requirements should be included in the written specification, initial qualification tests may only be needed prior to the first order. Routine production and receipt requirements should be adhered to on every order. Initial qualification requirements are indicated within each clause, where applicable.
This guide provides an understanding of the requirements needed for the manufacture, purchase, and acceptance of a preformed peelable pouch. Appropriate test methods for compliance are also cited.
Note 1—All test methods for a particular requirement may not be cited due to specific or unique circumstances. For additional guidance on applicable methods, refer to Guide F2097.
The specification and its requirements should be mutually agreed to by the supplier and purchaser of pouches. This helps ensure that pouches will comply to specified requirements.
1.1 This guide defines the requirements and considerations for flexible peel pouches with one open, unsealed end that are intended to be sterilized containing medical devices. These are also known as preformed sterile barrier systems.
1.2 Pouch styles are categorized as chevron, header, and corner peel. These pouches are typically manufactured by heat sealing, or in some cases, by cohesive cold sealing. The sealing bond is intended to be peeled open to aseptically dispense the contents.
1.3 Pouch materials may be either porous, nonporous, or any combination of the two.
1.4 This guide addresses some critical printing requirements on the pouch.
1.5 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E122 Practice for Calculating Sample Size to Estimate, With Specified Precision, the Average for a Characteristic of a Lot or Process
F17 Terminology Relating to Flexible Barrier Packaging
F88 Test Method for Seal Strength of Flexible Barrier Materials
F1140 Test Methods for Internal Pressurization Failure Resistance of Unrestrained Packages
F1886 Test Method for Determining Integrity of Seals for Medical Packaging by Visual Inspection
F2054 Test Method for Burst Testing of Flexible Package Seals Using Internal Air Pressurization Within Restraining Plates
F2097 Guide for Design and Evaluation of Primary Flexible Packaging for Medical Products
F2203 Test Method for Linear Measurement Using Precision Steel Rule
F2250 Practice for Evaluation of Chemical Resistance of Printed Inks and Coatings on Flexible Packaging Materials
F2475 Guide for Biocompatibility Evaluation of Medical Device Packaging Materials
TAPPI StandardsT437 Dirt in Paper and Paperboard
ICS Number Code 11.120.99 (Other standards related to pharmaceutics)