| ||Format||Pages||Price|| |
|6||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$44.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$52.80||  ADD TO CART|
Significance and Use
5.1 To be meaningful, walkway tribometer results must correlate the slip characteristics of a surface or contaminant, or both, to the actual propensity for human slips. To achieve this goal, walkway tribometer models must be validated against a standard with relevance to human ambulation.
5.2 This practice prescribes a series of reference surfaces with known relative slip potential ranging from very high to low (as defined by laboratory conditions only) upon which walkway tribometer models can be validated. The relative slip potential of each reference surface was established from human subject walking trials.3
5.3.1 The scientific study upon which the validation process is based was conducted with a select population of young adults (mean age 26 years) who were free from gait deviations while walking in a straight path on a level surface with a mean walking velocity of 2.18 m/s. This walking velocity is faster than the average walking velocity for the general population which includes a much wider age range with greater variability; thus, the study sample population of pedestrians and conditions is not representative of the larger general population of pedestrians.
5.3.2 All subjects walked in Oxford-style shoes whose soles were constructed of smooth styrene butadiene rubber (SBR) with 75A Shore hardness. The shoe style and sole material is not representative of all combinations available in the marketplace.
5.3.3 The reference surfaces defined in this practice are not representative of all walkway surfaces. The outcome of the validation practice reflects performance on the type of reference surfaces and surface conditions defined in this practice only. Validation and calibration of a walkway tribometer as defined by this practice does not imply validation and calibration under all combinations of test foot materials and walkway surfaces.
1.4 This practice provides a procedure through which an entity may certify a walkway tribometer model, signifying that the walkway tribometer model has a completed and documented validation and interlaboratory study.
1.5 This practice describes the necessary materials, specifications, and the cleaning process for reference materials, as well as the requirements for the validation of a supplier’s walkway tribometer and calibration of a user’s walkway tribometer.
1.6 This practice applies to walkway tribometers without reference to the nature of the scale of the readings produced by them. The scale used in the reports of validation and calibration must be the same, and are to be those of the instrument or defined for the instrument.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1349 Practice for Rubber--Standard Temperatures for Testing
D3244 Practice for Utilization of Test Data to Determine Conformance with Specifications
F1646 Terminology Relating to Safety and Traction for Footwear
ICS Number Code 17.040.20 (Properties of surfaces); 93.080.10 (Road construction)
UNSPSC Code 41113710(Level meter)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F2508-13, Standard Practice for Validation, Calibration, and Certification of Walkway Tribometers Using Reference Surfaces, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top