| ||Format||Pages||Price|| |
|13||$48.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$48.00||  ADD TO CART|
Significance and Use
4.2 This guide is intended for use by researchers and manufacturers for the development and selection of pre-test treatments, tests and test endpoints to measure stent securement (displacement distances and dislodgment forces).
4.4 This guide should be used with discretion in choosing securement tests and evaluating results due to the myriad possible combinations of clinical conditions, failure modes, and stent delivery system designs.
1.1 This guide provides guidance for the design and development of pre-test treatments, tests, and test endpoints to measure stent securement of pre-mounted, unsheathed, balloon-expandable stent delivery systems. This guide is intended to aid investigators in the design, development, and in vitro characterization of pre-mounted, unsheathed, balloon-expandable stent delivery systems.
1.2 This guide covers the laboratory determination of the shear force required to displace or dislodge a balloon-expandable endovascular stent mounted on a delivery system. The guide proposes a set of options to consider when testing stent securement. The options cover pre-test treatments, possible stent securement tests, and relevant test endpoints. An example test apparatus is given in 7.1.
1.3 This guide covers in vitro bench testing characterization only. Measured levels of securement and product design/process differentiation may be particularly influenced by selections of pre-test treatments, securement test type (for example, stent gripping method), and test endpoint. In vivo characteristics may also differ from in vitro results.
1.4 This guide does not cover all possible pre-test treatments, stent securement tests, or test endpoints. It is intended to provide a starting point from which to select and investigate securement test options.
1.6 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E1169 Practice for Conducting Ruggedness Tests
E1488 Guide for Statistical Procedures to Use in Developing and Applying Test Methods
ICS Number Code 11.040.20 (Transfusion, Infusion)
UNSPSC Code 42203401(Coronary stents)