| ||Format||Pages||Price|| |
|PDF Version||4||$37.00||  ADD TO CART|
|Print Version||4||$37.00||  ADD TO CART|
This specification covers unfabricated and fabricated forms of high purity hydrated calcium sulfate hemihydrate or dihydrate for surgical implants. The requirements of this specification apply to calcium sulfate combined with two molecules of water or two calcium sulfate molecules with one water mole but does not include calcium sulfate anhydrite and calcium sulfate forms that contain reinforcing phases, medicaments, biological agents, and other such additives. All covered materials should conform to the requirements for set time, compressive strength, and in vitro degradation.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers material requirements for unfabricated and fabricated forms of hydrated calcium sulfate intended for surgical implants. Fabricated forms may include pressed and cast surgical implants in various geometric shapes. The calcium sulfate hemihydrate in the unfabricated form can be converted with the addition of water or other water-containing solutions to a fabricated calcium sulfate dihydrate form.
1.2 The requirements of this specification apply to calcium sulfate combined with two molecules of water or two calcium sulfate molecules sharing one water molecule.
Approximate chemical formulae:
Calcium Sulfate Dihydrate
Calcium Sulfate Hemihydrate
CaSO4·1/2H2O or CaSO4·H2O·CaSO4
1.3 This specification specifically excludes calcium sulfate anhydrite and calcium sulfate forms that contain additives such as reinforcing phases, medicaments, biological agents, and so forth.
1.4 The presence of processing aids does not exclude a product from the physical and mechanical requirements of this specification.
1.5 Some provisions of Specification C59/C59M and Test Methods C472 apply. Special requirements that are detailed in this specification are included to characterize the material which will be used in surgical implants.
1.6 The biological response to calcium sulfate in bone tissue has been well characterized by a history of clinical use (1-14)2 and by laboratory studies (15-18).
1.7 The following precautionary caveat pertains only to the test method portion, Sections 4, 5, and 6, of this specification. This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory requirements prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C59/C59M Specification for Gypsum Casting Plaster and Gypsum Molding Plaster
C472 Test Methods for Physical Testing of Gypsum, Gypsum Plasters and Gypsum Concrete
F648 Specification for Ultra-High-Molecular-Weight Polyethylene Powder and Fabricated Form for Surgical Implants
F756 Practice for Assessment of Hemolytic Properties of Materials
F763 Practice for Short-Term Screening of Implant Materials
F813 Practice for Direct Contact Cell Culture Evaluation of Materials for Medical Devices
F895 Test Method for Agar Diffusion Cell Culture Screening for Cytotoxicity
F981 Practice for Assessment of Compatibility of Biomaterials for Surgical Implants with Respect to Effect of Materials on Muscle and Bone
F1088 Specification for Beta-Tricalcium Phosphate for Surgical Implantation
F1635 Test Method for in vitro Degradation Testing of Hydrolytically Degradable Polymer Resins and Fabricated Forms for Surgical Implants
ICS Number Code 11.040.40 (Implants for surgery, prothetics and orthotics)
UNSPSC Code 42295500(Surgical implants and expanders and extenders and surgical wires and related products)