Active Standard ASTM F2101 | Developed by Subcommittee: F23.40
Book of Standards Volume: 11.03
Historical (view previous versions of standard)
1.1 This test method is used to measure the bacterial filtration efficiency (BFE) of medical face mask materials, employing a ratio of the upstream bacterial challenge to downstream residual concentration to determine filtration efficiency of medical face mask materials.
1.2 This test method is a quantitative method that allows filtration efficiency for medical face mask materials to be determined. The maximum filtration efficiency that can be determined by this method is 99.9 %.
1.3 This test method does not apply to all forms or conditions of biological aerosol exposure. Users of the test method should review modes for worker exposure and assess the appropriateness of the method for their specific applications.
1.4 This test method evaluates medical face mask materials as an item of protective clothing but does not evaluate materials for regulatory approval as respirators. If respiratory protection for the wearer is needed, a NIOSH-certified respirator should be used. Relatively high bacterial filtration efficiency measurements for a particular medical face mask material does not ensure that the wearer will be protected from biological aerosols since this test method primarily evaluates the performance of the composite materials used in the construction of the medical face mask and not its design, fit or facial sealing properties.
1.5 Units The values stated in SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in nonconformance of the standard.
1.6 This test method does not address breathability of the medical face mask materials or any other properties affecting the ease of breathing through the medical face mask material.
This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E171 Practice for Conditioning and Testing Flexible Barrier Packaging
F1494 Terminology Relating to Protective Clothing
ANSI/ASQC StandardANSI/ASQCZ1.4 Sampling Procedures and Tables for Inspection by Attributes
ISO StandardISO2859-1 Sampling Plans for Inspection by Attributes
Military StandardMIL-STD36954C(1973) Military Specification: Mask, Surgical, Disposable
ICS Number Code 11.140 (Hospital equipment)
UNSPSC Code 46182001(Masks or accessories)