| ||Format||Pages||Price|| |
|6||$43.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$43.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$51.60||  ADD TO CART|
This specification covers the chemical, physical, mechanical, and metallurgical requirements for wrought nickel-titanium bar, flat rolled products, and tubing containing nominally 54.5 % to 57.0 % nickel and used for the manufacture of medical devices and surgical implants. The material shall be made from ingot made from nickel and titanium with no other intentional alloy additions. The material shall be vacuum or inert atmosphere melted to control metallurgical cleanliness and alloy chemistry. Bar, plate, and tubing shall be supplied as hot finished or cold finished and annealed or heat treated. Surface condition may be oxidized, descaled, pickled, blasted, machined, ground, mechanically polished, or electropolished. Major elements shall be analyzed by direct current plasma spectrometry, atomic absorption, inductively coupled plasma spectrometry, X-ray spectrometer, glow discharge mass spectrometry, or an equivalent method. Carbon shall be measured by combustion and hydrogen shall be measured by inert gas fusion or vacuum hot extraction. Nitrogen and oxygen shall be measured by inert gas fusion. The nickel and titanium contents of nickel-titanium shape memory alloys cannot be measured to a precision required to guarantee shape memory or superelastic properties. Calorimetry or an equivalent thermomechanical test method must be used to assure the alloy formulation in terms of transformation temperature. Mechanical properties of the samples such as tensile strength and elongation shall be determined after annealing.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers the chemical, physical, mechanical, and metallurgical requirements for wrought nickel-titanium bar, flat rolled products, and tubes containing nominally 54.5- to 57.0-weight percent nickel and used for the manufacture of medical devices and surgical implants.
1.2 Requirements are for mill product, measuring 5.50 to 94.0 mm [0.218 to 3.70 in.] diameter or thickness. Mill product is not intended to have the final shape, final surface finish, or final properties of the medical device, implant, or their components. Finished NiTi cold-worked tube should be considered under Specification F2633.
1.3 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E4 Practices for Force Verification of Testing Machines
E8/E8M Test Methods for Tension Testing of Metallic Materials
E29 Practice for Using Significant Digits in Test Data to Determine Conformance with Specifications
E112 Test Methods for Determining Average Grain Size
E1019 Test Methods for Determination of Carbon, Sulfur, Nitrogen, and Oxygen in Steel, Iron, Nickel, and Cobalt Alloys by Various Combustion and Fusion Techniques
E1097 Guide for Determination of Various Elements by Direct Current Plasma Atomic Emission Spectrometry
E1172 Practice for Describing and Specifying a Wavelength-Dispersive X-Ray Spectrometer
E1245 Practice for Determining the Inclusion or Second-Phase Constituent Content of Metals by Automatic Image Analysis
E1409 Test Method for Determination of Oxygen and Nitrogen in Titanium and Titanium Alloys by the Inert Gas Fusion Technique
E1447 Test Method for Determination of Hydrogen in Titanium and Titanium Alloys by Inert Gas Fusion Thermal Conductivity/Infrared Detection Method
E1479 Practice for Describing and Specifying Inductively-Coupled Plasma Atomic Emission Spectrometers
E1941 Test Method for Determination of Carbon in Refractory and Reactive Metals and Their Alloys by Combustion Analysis
F1710 Test Method for Trace Metallic Impurities in Electronic Grade Titanium by High Mass-Resolution Glow Discharge Mass Spectrometer
F2004 Test Method for Transformation Temperature of Nickel-Titanium Alloys by Thermal Analysis
F2005 Terminology for Nickel-Titanium Shape Memory Alloys
F2633 Specification for Wrought Seamless Nickel-Titanium Shape Memory Alloy Tube for Medical Devices and Surgical Implants
Other StandardsISO9001 Quality Management Systems--Requirements Available from American National Standards Institute (ANSI), 25 W. 43rd St., 4th Floor, New York, NY 10036, http://www.ansi.org.
ICS Number Code 11.040.30 (Surgical instruments and materials)
UNSPSC Code 42294802(Flexible endoscopes or accessories or related products)