| ||Format||Pages||Price|| |
|6||$43.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$43.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$51.60||  ADD TO CART|
Significance and Use
4.1 This practice is a guideline for a screening test for the evaluation of the local tissue response to materials that may be selected for implantation into the human body and which are expected to undergo absorption within three years.
4.2 This practice is similar to that for studies on candidate materials that are not absorbable, such as those specified in Practices , , and ; however, analysis of the host response must take into account the effect of degradation and degradation products on the inflammatory response at the local tissue site and on subsequent healing of the implantation site.
4.3 The material to be tested should be in the final finished form as for intended use, including sterilization. Material/body ratios should be relevant to that of intended device use. Material surface area or mass to body mass ratios of 1X, 10X, and 50X if applicable, are recommended.
4.4 Materials that are designed for use in devices with in situ polymerization shall be introduced in a manner such that in situ polymerization occurs. Testing of individual precursor components is not recommended.
1.1 This practice provides experimental protocols for biological assays of tissue reactions to absorbable biomaterials for implant applications. This practice applies only to absorbable materials with projected clinical applications in which the materials will reside in bone or soft tissue longer than 30 days and less than three years. Other standards with designated implantation times are available to address shorter time periods. Careful consideration should be given to the appropriateness of this practice for slowly degrading materials that will remain for longer than three years. It is anticipated that the tissue response to degrading biomaterials will be different from the response to nonabsorbable materials. In many cases, a chronic inflammatory response may be observed during the degradation phase, but the local histology should return to normal after absorption; therefore, the minimal tissue response usually equated with “biocompatibility” may require long implantations.
1.2 The time period for implant absorption will vary depending on chemical composition implant size, implant location, and test subject species; therefore, the implantation times for examination of tissue response will be linked to the rate of absorption. No single implantation time is indicated in this practice.
1.3 These protocols assess the effects of the material on the animal tissue in which it is implanted. The experimental protocols do not fully assess systemic toxicity, carcinogenicity, teratogenicity, or mutagenicity of the material. Other standards are available to address these issues.
1.4 To maximize use of the animals in the study protocol, all toxicological findings should be recorded. There are some aspects of systemic toxicity, including effects of degradation products on the target organs, that can be addressed with this practice, and these effects should be documented fully.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
F561 Practice for Retrieval and Analysis of Medical Devices, and Associated Tissues and Fluids
F750 Practice for Evaluating Material Extracts by Systemic Injection in the Mouse
F763 Practice for Short-Term Screening of Implant Materials
F981 Practice for Assessment of Compatibility of Biomaterials for Surgical Implants with Respect to Effect of Materials on Muscle and Bone
F1408 Practice for Subcutaneous Screening Test for Implant Materials
F1903 Practice for Testing For Biological Responses to Particles In Vitro
F1904 Practice for Testing the Biological Responses to Particles in vivo
F1905 Practice For Selecting Tests for Determining the Propensity of Materials to Cause Immunotoxicity
F1906 Practice for Evaluation of Immune Responses In Biocompatibility Testing Using ELISA Tests, Lymphocyte Proliferation, and Cell Migration
ICS Number Code 11.040.40 (Implants for surgery, prothetics and orthotics)
UNSPSC Code 42295500(Surgical implants and expanders and extenders and surgical wires and related products)
ASTM F1983-14, Standard Practice for Assessment of Selected Tissue Effects of Absorbable Biomaterials for Implant Applications, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top