Active Standard ASTM F1828 | Developed by Subcommittee: F04.34
Book of Standards Volume: 13.01
Historical (view previous versions of standard)
This specification covers the chemical, mechanical, and metallurgical requirements for wrought titanium-12 molybdenum- 6 zirconium-2 iron alloy for surgical implants to be used in the manufacture of surgical implants. The heat analysis shall conform to the chemical composition requirements prescribed. Ingot analysis may be used for reporting all chemical requirements, except hydrogen. The wrought titanium-12 molybdenum-6 zirconium-2 iron alloy are classified as bar, forging bar and wire. The ultimate tensile strength, yield strength, elongation, and area reduction of the material shall be tested to meet the requirements prescribed.This specification covers the referee test methods for evaluating the performance characteristics of a single-use ureteral stent with retaining means at both ends, during short term use for drainage of urine from the kidney to the bladder. Ureteral stents shall be tested in accordance with the appropriate biological tests to meet the requirements prescribed. Retention strength, break strength, elongation, dynamic frictional force, and radiopacity shall be tested to meet the requirements prescribed.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers the referee test methods for evaluating the performance characteristics of a single-use ureteral stent with retaining means at both ends, during short term use for drainage of urine from the kidney to the bladder. These stents are typically available in diameters of 3.7 Fr to 14.0 Fr, and lengths of 8 cm to 30 cm, and are made of silicone, polyurethane, and other polymers. They are provided non-sterile for sterilization and sterile for single-use.
1.2 Exclusions—Long-term indwelling usage (over 30 days) is encountered with this product, but not commonly, and is therefore considered an exception to this specification. Similarly, the use of ureteral stents for non-ureteral applications such as nephrostomy and ileostomy is excluded from the scope of this specification. Non-sterile ureteral stents are also excluded due to the variability of hospital sterilization equipment and processes and the resulting effects on ureteral stent characteristics.
1.3 The following precautionary statement pertains only to the test method portion, Section 5, of this specification:
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D412 Test Methods for Vulcanized Rubber and Thermoplastic Elastomers--Tension
F640 Test Methods for Determining Radiopacity for Medical Use
F748 Practice for Selecting Generic Biological Test Methods for Materials and Devices
ICS Number Code 11.040.40 (Implants for surgery, prothetics and orthotics)
UNSPSC Code 42295903(Urological stents)