| ||Format||Pages||Price|| |
|6||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$44.00||  ADD TO CART|
This specification covers portable, live-line tool-supported, direct-contact type capacitive voltage detectors to be used on electrical systems both indoors and outdoors for ac voltages. This specification establishes requirements for the design and testing of high voltage detectors, used in the electrical power industry, to determine the presence or absence of nominal operating voltage. The following tests shall be performed: voltage; low temperature impact; drop/impact; humidity; wet test; battery life test; durability of labeling; vibration resistance; continuous operation rating; response time; testing the self-test function; acceptable audible indication; acceptable visual indication; visual inspection; method to measure threshold voltage; interference voltage testing; leakage current testing; dielectric testing for detector housing; and wet testing.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
Significance and Use
4.1 This specification establishes requirements for the design and testing of high voltage detectors, used in the electrical power industry, to determine the presence or absence of nominal operating voltage or the measured voltage.
1.1 This specification covers portable, live-line tool-supported, direct-contact type capacitive voltage detectors to be used on electrical systems both indoors and outdoors for ac voltages from 600 V to 800 kV with frequency of 60 Hz. The function of the voltage detector is limited to the detection of the presence or absence of nominal operating voltage.
1.1.1 Two types of voltage detectors are provided and are designated as Type I, audible/visual and Type II, numeric, with or without audible.
1.1.2 Two styles of voltage detectors, differing in wet conditions characteristics, are provided and are designated as Style A, indoor use and Style B indoor/outdoor use.
1.2 The use and maintenance of these high voltage detectors and any necessary insulated tool handles are beyond the scope of this specification.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Note 1: Except where specified, all voltage defined in this specification refer to phase-to-phase voltage in a three-phase system. Voltage detectors covered by this specification may be used in other than three-phase systems, but the applicable phase-to-phase or phase-to-ground (earth) voltages shall be used to determine the operating voltage.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
F819 Terminology Relating to Electrical Protective Equipment for Workers
ANSI StandardANSI C84.1 Voltage Ratings for Electric Power Systems and Equipment
ICS Number Code 17.220.20 (Measurement of electrical and magnetic quantities)
UNSPSC Code 32101519(Detectors)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F1796-09(2014), Standard Specification for High Voltage Detectors—Part 1 Capacitive Type to be Used for Voltages Exceeding 600 Volts AC, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top