Significance and Use
Chloride ion is highly detrimental to the valves, flanges, piping, etc. of stainless steel and copper-bearing alloys employed in certain types of facilities, such as nuclear and chemical plants. Therefore, the amount of chloride ion that can be leached from packing and gasketing materials used in these facilities must be controlled closely for prevention of damage. This test method is suitable for manufacturing control and for verifying that the packing and gasketing material meets specifications.
It is assumed that the users of this test method will be analysts capable of performing common laboratory procedures skillfully and safely. It is recommended that the work will be performed in a properly equipped laboratory under appropriate quality control practices such as those described in Guide E882.
1.1 This test method provides for the measurement of chloride ion leached from flexible graphite, asbestos, or paper-based packing and gasketing materials. Samples containing 7 to 6700 μg Cl-/g sample can be analyzed with this procedure. The upper concentration limit can be extended by dilution of the test solution.
1.2 It is the responsibility of the analyst to ensure the validity of this test method for other packing or gasketing materials.
1.3 The values stated in SI units are to be regarded as the standard. The values given in parentheses are for information only.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. A specific warning statement is given in 8.3.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D512 Test Methods for Chloride Ion In Water
E50 Practices for Apparatus, Reagents, and Safety Considerations for Chemical Analysis of Metals, Ores, and Related Materials
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E882 Guide for Accountability and Quality Control in the Chemical Analysis Laboratory
ion-selective electrode technique; leachable chloride; packing and gasketing materials; Asbestos; Chloride analysis; Damage assessment; Gaskets; Graphite; Ion selective electrode method; Manufacturing processes; Packing/packaging materials/systems; Paper-based gasket materials; Potentiometric method; Quality control (QC) ;
ICS Number Code 91.100.50 (Binders. Sealing materials)
ASTM International is a member of CrossRef.
Citing ASTM Standards
[Back to Top]