| ||Format||Pages||Price|| |
|5||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$44.00||  ADD TO CART|
|Standard + Redline PDF Bundle||10||$52.80||  ADD TO CART|
Significance and Use
5.1 The calculated error in the photovoltaic device current determined from the spectral mismatch parameter can be used to determine if a measurement will be within specified limits before the actual measurement is performed.
5.2 The spectral mismatch parameter also provides a means of correcting the error in the measured device current due to spectral mismatch.
5.2.1 The spectral mismatch parameter is formulated as the fractional error in the short-circuit current due to spectral and temperature differences.
5.2.2 Error due to spectral mismatch is corrected by multiplying a reference cell’s measured short-circuit current by M , a technique used in Test Methods and .
5.3 Because all spectral quantities appear in both the numerator and the denominator in the calculation of the spectral mismatch parameter (see ), multiplicative calibration errors cancel, and therefore only relative quantities are needed (although absolute spectral quantities may be used if available).
5.4 Temperature-dependent spectral mismatch is a more accurate method to correct photovoltaic current measurements compared with fixed-value temperature coefficients.
1.1 This test method provides a procedure for the determination of a spectral mismatch parameter used in performance testing of photovoltaic devices.
1.2 The spectral mismatch parameter is a measure of the error introduced in the testing of a photovoltaic device that is caused by the photovoltaic device under test and the photovoltaic reference cell having non-identical quantum efficiencies, as well as mismatch between the test light source and the reference spectral irradiance distribution to which the photovoltaic reference cell was calibrated.
1.2.1 Examples of reference spectral irradiance distributions are Tables or .
1.3 The spectral mismatch parameter can be used to correct photovoltaic performance data for spectral mismatch error.
1.4 Temperature-dependent quantum efficiencies are used to quantify the effects of temperature differences between test conditions and reporting conditions.
1.5 This test method is intended for use with linear photovoltaic devices in which short-circuit is directly proportional to incident irradiance.
1.6 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E490 Standard Solar Constant and Zero Air Mass Solar Spectral Irradiance Tables
E772 Terminology of Solar Energy Conversion
E948 Test Method for Electrical Performance of Photovoltaic Cells Using Reference Cells Under Simulated Sunlight
E1021 Test Method for Spectral Responsivity Measurements of Photovoltaic Devices
E1036 Test Methods for Electrical Performance of Nonconcentrator Terrestrial Photovoltaic Modules and Arrays Using Reference Cells
E1039 Test Method for Calibration of Silicon Non-Concentrator Photovoltaic Primary Reference Cells Under Global Irradiation
E1125 Test Method for Calibration of Primary Non-Concentrator Terrestrial Photovoltaic Reference Cells Using a Tabular Spectrum
E1328 Terminology Relating to Photovoltaic Solar Energy Conversion
ICS Number Code 27.160 (Solar energy engineering)
UNSPSC Code 32111701(Photovoltaic cells)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E973-15, Standard Test Method for Determination of the Spectral Mismatch Parameter Between a Photovoltaic Device and a Photovoltaic Reference Cell , ASTM International, West Conshohocken, PA, 2015, www.astm.orgBack to Top