Significance and Use
Metals and other materials are not always isotropic in their physical properties. For example, Young's modulus will vary in different crystallographic directions. Therefore, it is desirable or necessary to determine the orientation of a single crystal undergoing tests in order to ascertain the relation of any property to different directions in the material.
This test method can be used commercially as a quality control test in production situations where a desired orientation, within prescribed limits, is required.
With the use of an adjustable fixed holder that can later be mounted on a saw, lathe, or other machine, a single crystal material can be moved to a preferred orientation, and subsequently sectioned, ground, or processed otherwise.
If grains of a polycrystalline material are large enough, this test method can be used to determine their orientations and differences in orientation.
1.1 This test method covers the back-reflection Laue procedure for determining the orientation of a metal crystal. The back-reflection Laue method for determining crystal orientation (1, 2) may be applied to macrograins (3) (0.5-mm diameter or larger) within polycrystalline aggregates, as well as to single crystals of any size. The method is described with reference to cubic crystals; it can be applied equally well to hexagonal, tetragonal, or orthorhombic crystals.
1.2 Most natural crystals have well developed external faces, and the orientation of such crystals can usually be determined from inspection. The orientation of a crystal having poorly developed faces, or no faces at all (for example, a metal crystal prepared in the laboratory) must be determined by more elaborate methods. The most convenient and accurate of these involves the use of X-ray diffraction. The “orientation of a metal crystal” is known when the positions in space of the crystallographic axes of the unit cell have been located with reference to the surface geometry of the crystal specimen. This relation between unit cell position and surface geometry is most conveniently expressed by stereographic or gnomonic projection.
1.3 The values stated in inch-pound units are to be regarded as standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E3 Guide for Preparation of Metallographic Specimens
back reflection Laue X-ray; hyperbolic chart; metal crystal; orientation; polar chart; standard projection; X-ray diffraction; Back-reflection Laue method; Crystallographic characteristics; Laue method; Metal crystals; Nondestructive evaluation (NDE)--Laue method; Orientation; X-ray diffraction analysis;
ICS Number Code 77.040.99 (Other methods of testing metals)
ASTM International is a member of CrossRef.
Citing ASTM Standards
[Back to Top]