Active Standard ASTM E788 | Developed by Subcommittee: E41.01
Book of Standards Volume: 14.04
Historical (view previous versions of standard)
This specification covers the requirements for glass reusable blood diluting pipets that are used for performing red and white cell corpuscle determinations. The pipets shall be made of common spirit bore white back tubing or clear glass with a white stripe applied to the outer surface of the tubing. The beads shall be made of glass or ceramic composition and shall be of red, clear, or white coloring. Materials shall be tested and the individual types shall conform to specified values of design, capacity, capacity markings, graduation lines, accuracy, workmanship, identification, and pigmentation.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers requirements for glass reusable blood diluting pipets that are used for performing red and white cell corpuscle determinations.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E694 Specification for Laboratory Glass Volumetric Apparatus
E920 Specification for Commercially Packaged Laboratory Apparatus
E921 Specification for Export Packaged Laboratory Apparatus
E1133 Practice for Performance Testing of Packaged Laboratory Apparatus for United States Government Procurements
E1157 Specification for Sampling and Testing of Reusable Laboratory Glassware
ICS Number Code 11.100 (Laboratory medicine)
UNSPSC Code 41121511(Serological pipettes)