| ||Format||Pages||Price|| |
|7||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||7||$44.00||  ADD TO CART|
Significance and Use
5.1 This test method may be used to determine and validate the performance of a particular thermomechanical analyzer apparatus.
5.2 This test method may be used to determine and validate the performance of a particular method based upon thermomechanical analyzer temperature or length change measurements.
5.3 This test method may be used to determine the repeatability of a particular apparatus, operator, or laboratory.
5.4 This test method may be used for specification and regulatory compliance purposes.
1.1 This test method provides procedures for validating temperature and length change measurements of thermomechanical analyzers (TMA) and analytical methods based upon the measurement of temperature and length change. Performance parameters include temperature repeatability, linearity and bias; and dimension change repeatability, detection limit, quantitation limit, linearity and bias.
1.2 Validation of apparatus performance and analytical methods is a necessary requirement for quality initiatives. Results may also be used for legal purposes.
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
ICS Number Code 17.200.20 (Temperature-measuring instruments)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2918-13, Standard Test Method for Performance Validation of Thermomechanical Analyzers, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top