Significance and Use
IIT Instruments are used to quantitatively measure various mechanical properties of thin coatings and other volumes of material when other traditional methods of determining material properties cannot be used due to the size or condition of the sample. This practice will establish the basic requirements for those instruments. It is intended that IIT based test methods will be able to refer to this practice for the basic requirements for force and displacement accuracy, reproducibility, verification, reporting, etc., that are necessary for obtaining meaningful test results.
IIT is not restricted to specific test forces, displacement ranges, or indenter types. This practice covers the requirements for a wide range of nano, micro, and macro (see ISO 14577-1) indentation testing applications. The various IIT instruments are required to adhere to the requirements of the practice within the their specific design ranges.
1.1 This practice defines the basic steps of Instrumented Indentation Testing (IIT) and establishes the requirements, accuracies, and capabilities needed by an instrument to successfully perform the test and produce the data that can be used for the determination of indentation hardness and other material characteristics. IIT is a mechanical test that measures the response of a material to the imposed stress and strain of a shaped indenter by forcing the indenter into a material and monitoring the force on, and displacement of, the indenter as a function of time during the full loading-unloading test cycle.
1.2 The operational features of an IIT instrument, as well as requirements for Instrument Verification Annex A1), Standardized Reference Blocks (Annex A2) and Indenter Requirements (Annex A3) are defined. This practice is not intended to be a complete purchase specification for an IIT instrument.
1.3 With the exception of the non-mandatory Appendix X4, this practice does not define the analysis necessary to determine material properties. That analysis is left for other test methods. Appendix X4 includes some basic analysis techniques to allow for the indirect performance verification of an IIT instrument by using test blocks.
1.4 Zero point determination, instrument compliance determination and the indirect determination of an indenters area function are important parts of the IIT process. The practice defines the requirements for these items and includes non-mandatory appendixes to help the user define them.
1.5 The use of deliberate lateral displacements is not included in this practice (that is, scratch testing).
1.6 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E3 Guide for Preparation of Metallographic Specimens
E74 Practice of Calibration of Force-Measuring Instruments for Verifying the Force Indication of Testing Machines
E92 Test Method for Vickers Hardness of Metallic Materials
E384 Test Method for Knoop and Vickers Hardness of Materials
E1875 Test Method for Dynamic Youngs Modulus, Shear Modulus, and Poissons Ratio by Sonic Resonance
E1876 Test Method for Dynamic Youngs Modulus, Shear Modulus, and Poissons Ratio by Impulse Excitation of Vibration
American Bearings Manufacturer Association Standard
ABMA10-1988 Metal Balls Available from American Bearing Manufacturers Association (ABMA), 2025 M Street, NW Suite 800 Washington, DC 20036, http://www.abma-dc.org.
ISO376 Metallic Materials--Calibration of Force-Proving Instruments for the Verification of Uniaxial Testing Machines
force displacement curve; indentation hardness; indentation modulus; indenter shape function; instrument compliance; instrumented indentation; zero point;
ICS Number Code 19.100 (Non-destructive testing)
ASTM International is a member of CrossRef.
Citing ASTM Standards
[Back to Top]