| ||Format||Pages||Price|| |
|4||$39.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$39.00||  ADD TO CART|
|Standard + Redline PDF Bundle||8||$46.80||  ADD TO CART|
Significance and Use
5.1 Rheological properties such as viscosity and storage and loss modulus change rapidly with temperature. High quality determinations of these properties depend upon a stable and well-known temperature of the measuring apparatus.
1.1 This test method describes the temperature calibration or conformance of rheometers. The applicable temperature range is 0 to 80°C however other ranges may be selected for the purpose at hand.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 There are no ISO equivalents to this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E473 Terminology Relating to Thermal Analysis and Rheology
E1142 Terminology Relating to Thermophysical Properties
ICS Number Code 17.060 (Measurement of volume, mass, density, viscosity)
UNSPSC Code 41112502(Rheometers)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2509-14, Standard Test Method for Temperature Calibration of Rheometers in Isothermal Mode, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top