ASTM E2503 - 13
Standard Practice for Qualification of Basket and Paddle Dissolution Apparatus
Significance and Use
2.1 This practice outlines a procedure for the mechanical calibration of paddle and basket dissolution units to ensure reproducibility of results.
2.2 Once a unit meets all of the mechanical specifications included in this practice, it is considered calibrated and further calibration with dissolution calibrator tablets is not required.
1.1 This practice covers the set-up and calibration of the paddle and basket dissolution apparatus.
1.2 Use of this practice may be applied to apparatus that have been modified to enable automatic dissolution testing (that is, a valve in the bottom of the vessel or sampling through the shaft).
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
ICS Number Code 11.120.99 (Other standards related to pharmaceutics)
UNSPSC Code 73161510(Chemical or pharmaceutical machinery or equipment manufacture services)
ASTM International is a member of CrossRef.
Back to Top