| ||Format||Pages||Price|| |
|10||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||10||$44.00||  ADD TO CART|
|Standard + Redline PDF Bundle||20||$52.80||  ADD TO CART|
Significance and Use
4.1 This test method provides for the quantitative determination of inorganic chloride (chloride ion) in monoethylene glycol (MEG), diethylene glycol (DEG) and triethylene glycol (TEG) using ion chromatography with conductivity detection. The analysis time is less than 5 min with little or no sample preparation required. Conductivity detection is a universal detection mode and is linear over the range of the method. Acceptable levels of chloride in polyester-grade and low-conductivity-grade MEG vary with the manufacturer’s specifications but are normally in the low mg/kg range. Knowledge of the chloride content in polyester-grade and low-conductivity-grade MEG is required to establish whether the MEG product meets specification requirements.
4.2 Glycols have high viscosities and a dilution with high quality deionized water may be required depending on the capability of the autosampler, if used, to deliver the injection. All standards and samples, whether diluted or not should be treated in the same manner.
1.1 This test method covers the determination of inorganic chloride (chloride ion) in monoethylene glycol (MEG), diethylene glycol (DEG) and triethylene glycol (TEG) in the range of 0.01 to 1.0 mg/kg by ion chromatography (IC).
1.2 Ethylene glycol can be analyzed directly by this test method without any sample preparation or diluted with high quality deionized water if an autosampler is used and dilution is necessary (that is, 50:50 or other suitable ratio).
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3.1 The exception is the additional information of (psi) in , , and .
1.4 Review the current Safety Data Sheets (SDS) for detailed information concerning toxicity, first-aid procedures and safety precautions.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and to determine the applicability of regulatory limitations prior to use. For specific hazard statements, see Section .
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1193 Specification for Reagent Water
E180 Practice for Determining the Precision of ASTM Methods for Analysis and Testing of Industrial and Specialty Chemicals
E300 Practice for Sampling Industrial Chemicals
ICS Number Code 71.040.40 (Chemical analysis)
UNSPSC Code 12352121(Organic hydroxides)
ASTM E2469-16, Standard Test Method for Chloride in Mono-, Di- and Tri-ethylene Glycol by Ion Chromatography, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top