WITHDRAWN, REPLACED BY E2933
| ||Format||Pages||Price|| |
|11||$60.00||  ADD TO CART|
Formerly under the jurisdiction of Committee E54 on Homeland Security Applications, this specification was withdrawn in February 2014 and replaced by Specification E2885 on Handheld Point Chemical Vapor Detectors (HPCVD) for Homeland Security Applications and Specification E2933 on Stationary Point Chemical Vapor Detectors (SPCVD) for Homeland Security Applications.
1.1 This specification covers the technical and mission requirements for the use of a CWVD and relates each of the performance and electrical shock and fire parameters to a detector requirement. Refer to Table 1 for reference material correlation.
1.2 This specification also defines the interfaces between the CWVD, communication systems, service platforms, and power sources. Refer to Table 1 for reference material correlation.
1.3 The CWVD will be used to sample air and report concentrations of the following nerve and blister CWAs: GA, GB, GD, GF, VX, HD, L, and HN
1.4 The definitions in this specification apply only to this specification and shall be the determining factors(s) when interpreting any word or combination(s) of words. Definitions for selected terms are in Section 3. The reader is strongly encouraged to review these definitions before reading the requirements section.
1.5 The values given in SI units are to be considered the standard. The values given in parentheses are for information only.
This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and to determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D543 Practices for Evaluating the Resistance of Plastics to Chemical Reagents
D2240 Test Method for Rubber Property--Durometer Hardness
Federal StandardsOSHA Technical Manual, Section VIII: Chapter 1 Occupational Safety Health Administration Technical Manual - Chemical Protective Clothing Available from Occupational Safety and Health Administration (OSHA), 200 Constitution Ave., NW, Washington, DC 20210, http://www.osha.gov.
Industry Standards00-LC-S0322 NRC Materials License Requirements The interested party may use, with government approval, standards, specifications, or handbooks that are technically equivalent or superior to the standards listed above. In the event of a conflict between the text of this specification and the references cited, this specification takes precedence. Nothing in this specification, however, supersedes applicable laws and regulations unless a specific exemption has been obtained. Any deviation from this standard shall be noted in the documentation certifying compliance.
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2411-07, Standard Specification for Chemical Warfare Vapor Detector (CWVD), ASTM International, West Conshohocken, PA, 2007, www.astm.orgBack to Top