Active Standard ASTM E1982 | Developed by Subcommittee: E13.03
Book of Standards Volume: 03.06
Historical (view previous versions of standard)
Significance and Use
4.1 An OP/FT-IR monitor can, in principle, measure the concentrations of all IR-active gases and vapors in the atmosphere. Detailed descriptions of OP/FT-IR systems and the fundamental aspects of their operation are given in Guide E1685 and the FT-IR Open-Path Monitoring Guidance Document. A method for processing OP/FT-IR data to obtain the concentrations of gases over a long, open path is given in Compendium Method TO-16. Applications of OP/FT-IR systems include monitoring for gases and vapors in ambient air, along the perimeter of an industrial facility, at hazardous waste sites and landfills, in response to accidental chemical spills or releases, and in workplace environments.
1.1 This practice covers procedures for using active open-path Fourier transform infrared (OP/FT-IR) monitors to measure the concentrations of gases and vapors in air. Procedures for choosing the instrumental parameters, initially operating the instrument, addressing logistical concerns, making ancillary measurements, selecting the monitoring path, acquiring data, analyzing the data, and performing quality control on the data are given. Because the logistics and data quality objectives of each OP/FT-IR monitoring program will be unique, standardized procedures for measuring the concentrations of specific gases are not explicitly set forth in this practice. Instead, general procedures that are applicable to all IR-active gases and vapors are described. These procedures can be used to develop standard operating procedures for specific OP/FT-IR monitoring applications.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E131 Terminology Relating to Molecular Spectroscopy
E168 Practices for General Techniques of Infrared Quantitative Analysis
E1421 Practice for Describing and Measuring Performance of Fourier Transform Mid-Infrared (FT-MIR) Spectrometers: Level Zero and Level One Tests
E1655 Practices for Infrared Multivariate Quantitative Analysis
E1685 Practice for Measuring the Change in Length of Fasteners Using the Ultrasonic Pulse-Echo Technique
Other DocumentsCompendiumMethodTO-1 Long-Path Open-Path Fourier Transform Infrared Monitoring of Atmospheric Gases Compendium of Methods for the Determination of Toxic Organic Compounds in Ambient Air, 2nd Ed., EPA/625/R-96/010b, Center for Environmental Research Info., Office of Research & Development, U.S. Environmental Protection Agency, Cincinnati, OH 45268, Jan. 1997.
ICS Number Code 17.180.30 (Optical measuring instruments)