| ||Format||Pages||Price|| |
|4||$38.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$38.00||  ADD TO CART|
|Standard + Redline PDF Bundle||8||$45.60||  ADD TO CART|
Significance and Use
4.1 CT may be performed on an object when it is in the as-cast, intermediate, or final machined condition. A CT examination can be used as a design tool to improve wax forms and moldings, establish process parameters, randomly check process control, perform final quality control (QC) examination for the acceptance or rejection of parts, and analyze failures and extend component lifetimes.
4.2 The most common applications of CT for castings are for the following: locating and characterizing discontinuities, such as porosity, inclusions, cracks, and shrink; measuring as-cast part dimensions for comparison with design dimensions; and extracting dimensional measurements for reverse engineering.
4.3 The extent to which a CT image reproduces an object or a feature within an object is dictated largely by the competing influences of spatial resolution, contrast discrimination, the specific geometry and material of the object itself, and artifacts of the imaging system. Operating parameters strike an overall balance between image quality, examination time, and cost.
4.4 Artifacts are often the limiting factor in CT image quality. (See Practice E1570 for an in-depth discussion of artifacts.) Artifacts are reproducible features in an image that are not related to actual features in the object. Artifacts can be considered correlated noise because they form repeatable fixed patterns under given conditions yet carry no object information. For castings, it is imperative to recognize what is and is not an artifact since an artifact can obscure or masquerade as a discontinuity. Artifacts are most prevalent in castings with long straight edges or complex geometries, or both.
1.1 This practice covers a uniform procedure for the examination of castings by the computed tomography (CT) technique. The requirements expressed in this practice are intended to control the quality of the nondestructive examination by CT and are not intended for controlling the acceptability or quality of the castings. This practice implicitly suggests the use of penetrating radiation, specifically X rays and gamma rays.
1.2 This practice provides a uniform procedure for a CT examination of castings for one or more of the following purposes:
1.2.1 Examining for discontinuities, such as porosity, inclusions, cracks, and shrink;
1.2.2 Performing metrological measurements and determining dimensional conformance; and
1.2.3 Determining reverse engineering data, that is, creating computer-aided design (CAD) data files.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E543 Specification for Agencies Performing Nondestructive Testing
E1316 Terminology for Nondestructive Examinations
E1441 Guide for Computed Tomography (CT) Imaging
E1570 Practice for Computed Tomographic (CT) Examination
E1672 Guide for Computed Tomography (CT) System Selection
E1695 Test Method for Measurement of Computed Tomography (CT) System Performance
E1935 Test Method for Calibrating and Measuring CT Density
E2339 Practice for Digital Imaging and Communication in Nondestructive Evaluation (DICONDE)
E2767 Practice for Digital Imaging and Communication in Nondestructive Evaluation (DICONDE) for X-ray Computed Tomography (CT) Test Methods
ASNT DocumentsANSI/ASNT CP-189 Standard for Personnel Qualification and Certification of Nondestructive Testing Personnel SNT-TC-1A Recommended Practice for Personnel Qualification and Certification in Nondestructive Testing
Military StandardsNAS 410 Certification and Qualification of Nondestructive Test Personnel
ICS Number Code 77.040.20 (Non-destructive testing of metals)
UNSPSC Code 42241600(Casting equipment and parts and accessories)
ASTM E1814-14, Standard Practice for Computed Tomographic (CT) Examination of Castings, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top