| ||Format||Pages||Price|| |
|3||$38.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$38.00||  ADD TO CART|
|Standard + Redline PDF Bundle||6||$45.60||  ADD TO CART|
Significance and Use
This guide is intended for use in evaluating the performance of field-portable electroanalytical or spectrophotometric devices for lead determination, or both.
Desired performance criteria for field-based extraction procedures are provided.
Performance parameters of concern may be determined using protocols that are referenced in this guide.
Example reference materials to be used in assessing the performance of field-portable lead analyzers are listed.
Exhaustive details regarding quality assurance issues are outside the scope of this guide. Applicable quality assurance aspects are dealt with extensively in references that are cited in this guide.
1.1 This standard provides guidelines for determining the performance of field-portable quantitative lead analysis instruments.
1.2 This guide applies to field-portable electroanalytical and spectrophotometric (including reflectance and colorimetric) analyzers.
1.3 Sample matrices of concern herein include paint, dust, soil, and airborne particles.
1.4 This guide addresses the desired performance characteristics of field-based sample extraction procedures for lead, as well as on-site extraction followed by field-portable analysis.
This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D5438 Practice for Collection of Floor Dust for Chemical Analysis
D6785 Test Method for Determination of Lead in Workplace Air Using Flame or Graphite Furnace Atomic Absorption Spectrometry
D6966 Practice for Collection of Settled Dust Samples Using Wipe Sampling Methods for Subsequent Determination of Metals
D7035 Test Method for Determination of Metals and Metalloids in Airborne Particulate Matter by Inductively Coupled Plasma Atomic Emission Spectrometry (ICP-AES)
D7144 Practice for Collection of Surface Dust by Micro-vacuum Sampling for Subsequent Metals Determination
E1605 Terminology Relating to Lead in Buildings
E1613 Test Method for Determination of Lead by Inductively Coupled Plasma Atomic Emission Spectrometry (ICP-AES), Flame Atomic Absorption Spectrometry (FAAS), or Graphite Furnace Atomic Absorption Spectrometry (GFAAS) Techniques
E1644 Practice for Hot Plate Digestion of Dust Wipe Samples for the Determination of Lead
E1645 Practice for Preparation of Dried Paint Samples by Hotplate or Microwave Digestion for Subsequent Lead Analysis
E1726 Practice for Preparation of Soil Samples by Hotplate Digestion for Subsequent Lead Analysis
E1727 Practice for Field Collection of Soil Samples for Subsequent Lead Determination
E1729 Practice for Field Collection of Dried Paint Samples for Subsequent Lead Determination
E1792 Specification for Wipe Sampling Materials for Lead in Surface Dust
E1864 Practice for Evaluating Quality Systems of Organizations Conducting Facility and Hazard Assessments for Lead in Paint, Dust, Airborne Particulate, and Soil in and around Buildings and Related Structures
U.S. EPA DocumentsEPA 747-R-92-001 Laboratory Accreditation Guidelines: Measurement of Lead in Paint, Dust, and Soil (1992)
ICS Number Code 71.040.50 (Physicochemical methods of analysis)
UNSPSC Code 11101714(Lead); 41115406(Spectrophotometers)
ASTM E1775-07, Standard Guide for Evaluating Performance of On-Site Extraction and Field-Portable Electrochemical or Spectrophotometric Analysis for Lead, ASTM International, West Conshohocken, PA, 2007, www.astm.orgBack to Top