Significance and Use
This practice is intended to describe the performance of a conductivity detector independent of the chromatographic system in terms that the analyst can use to predict overall system performance when the detector is coupled to the column and other chromatography system components.
Although it is possible to observe each of the several characteristics of a detector under different and unique conditions, it is the intent of this practice that a complete set of detector specifications should be obtained at the same operating conditions, including the setup used for testing, flow rates, and temperatures. It should be noted that to specify a detector's capability completely, its performance should be measured at several sets of conditions within the useful range of the detector. The terms and tests described in this practice are sufficiently general so that they may be used at whatever conditions may be chosen for other reasons.
1.1 This practice covers the testing of the performance of conductivity detectors used as the detection component of a liquid or ion chromatography system.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E1151 Practice for Ion Chromatography Terms and Relationships
conductivity detector; detector drift; detector flow-rate dependence; detector linear range; detector linearity; detector noise; detector response time; IC; ion chromatography; Liquid chromatography (LC); Conductivity detectors/detection; Detector linearity; Ion chromatography (IC); Ion exchange chromatography; Linear range;
ICS Number Code 71.040.50 (Physicochemical methods of analysis)
ASTM International is a member of CrossRef.
Citing ASTM Standards
[Back to Top]