Active Standard ASTM E1426 | Developed by Subcommittee: E28.13
Book of Standards Volume: 03.01
Historical (view previous versions of standard)
Significance and Use
This test method provides standard procedures for experimentally determining the effective elastic parameter for X-ray diffraction measurement of residual and applied stresses. It also provides a standard means of reporting the precision of the parameter.
This test method is applicable to any crystalline material which exhibits a linear relationship between stress and strain in the elastic range.
This test method should be used whenever residual stresses are to be evaluated by an X-ray diffraction technique and the effective elastic parameter of the material is unknown.
1.1 This test method covers a procedure for experimentally determining the effective elastic parameter, Eeff, for the evaluation of residual and applied stresses by X-ray diffraction techniques. The effective elastic parameter relates macroscopic stress to the strain measured in a particular crystallographic direction in polycrystalline samples. Eeff should not be confused with E, the modulus of elasticity. Rather, it is nominally equivalent to E/(1 + ν) for the particular crystallographic direction, where ν is Poisson's ratio. The effective elastic parameter is influenced by elastic anisotropy and preferred orientation of the sample material.
1.2 This test method is applicable to all X-ray diffraction instruments intended for measurements of macroscopic residual stress that use measurements of the positions of the diffraction peaks in the high back-reflection region to determine changes in lattice spacing.
1.3 This test method is applicable to all X-ray diffraction techniques for residual stress measurement, including single, double, and multiple exposure techniques.
1.4 The values stated in inch-pound units are to be regarded as standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E4 Practices for Force Verification of Testing Machines
E6 Terminology Relating to Methods of Mechanical Testing
E7 Terminology Relating to Metallography
E1237 Guide for Installing Bonded Resistance Strain Gages
ICS Number Code 71.040.50 (Physicochemical methods of analysis)