| ||Format||Pages||Price|| |
|6||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$44.00||  ADD TO CART|
Significance and Use
5.1 This test method should enable the user of the balance to interpret data determined thereon in terms of accuracy and precision. It should be helpful in using a particular instrument to best advantage. Weaknesses as well as strengths should become apparent. It is not the intention of this test method to compare similar instruments of different manufacture but rather to assist in choosing an instrument which will meet the needs of the user.
1.1 This test method can be used for testing equal-arm balances of any capacity and sensitivity. The testing procedure should enable the user to characterize his instrument sufficiently to determine whether or not it is suitable for the purpose for which it is to be used.
1.2 The characteristics to be examined include:
1.2.1 Sensitivity at all loads,
1.2.2 Lever arm ratio,
1.2.3 Damping ratio (for instruments without accessory dampers),
1.2.4 Period of oscillation,
1.2.5 Precision, and
1.2.6 Linearity and calibration of accessory devices that provide on-scale indication of weight.
1.3 This standard does not purport to address all of the safety concerns associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E617 Specification for Laboratory Weights and Precision Mass Standards
UNSPSC Code 41104000(Sampling equipment)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E1270-88(2014), Standard Test Method for Equal Arm Balances, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top