| ||Format||Pages||Price|| |
|3||$39.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$39.00||  ADD TO CART|
|Standard + Redline PDF Bundle||6||$46.80||  ADD TO CART|
This specification states the requirements for sound sources used for measuring the speech privacy between open offices or for measuring the laboratory performance of acoustical components. The sound source shall be a loudspeaker enclosed in a box that has a maximum dimension of 0.30 m (1 ft) on a side, to reduce spurious sound reflections. The measurements shall be carried out in a free sound field. The measurement microphone, amplifier, and level meter used to measure sound pressure levels shall satisfy the requirements prescribed. When the sound source is driven with the qualification signal, the sound output shall be adequate to maintain one-third octave-band sound pressure levels at least 10 dB above the corresponding background noise in each band at each measurement location. The directivity of the sound source shall be verified by driving the source with the qualification signal and measuring the sound pressure levels at measurement points.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification states the requirements for sound sources used for measuring the speech privacy between open offices and for measuring the laboratory performance of acoustical components (see Test Methods E1111 and E1130).
1.2 The sound source shall be a loudspeaker located in an enclosure driven with an appropriate test signal.
1.3 This specification describes the sound source and method of qualifying it using a special qualification signal. Test signals required by open office test methods may differ.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C384 Test Method for Impedance and Absorption of Acoustical Materials by Impedance Tube Method
C634 Terminology Relating to Building and Environmental Acoustics
E1050 Test Method for Impedance and Absorption of Acoustical Materials Using a Tube, Two Microphones and a Digital Frequency Analysis System
E1111 Test Method for Measuring the Interzone Attenuation of Open Office Components
E1130 Test Method for Objective Measurement of Speech Privacy in Open Plan Spaces Using Articulation Index
ANSI StandardsS1.11 Specification for Octave Band and Fractional OB Analog and Digital Filters S1.4 Specification for Sound Level Meters S1.43 Specifications for Integrating-Averaging Sound Level Meters S1.6 Preferred Frequencies, Frequency Levels, and Band Numbers for Acoustical Measurements
ICS Number Code 17.140.01 (Acoustic measurements and noise abatement in general)
UNSPSC Code 41115503(Sound measuring apparatus or decibel meter)
ASTM E1179-13, Standard Specification for Sound Sources Used for Testing Open Office Components and Systems, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top