| ||Format||Pages||Price|| |
|3||$37.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$37.00||  ADD TO CART|
This specification covers non-insulated platinum rhodium alloys (weight percent composition), and platinum thermoelements. The purity of platinum for use in the negative thermo element of Type R and Type S thermocouples shall be of a specific Grade, as specified. The purity of platinum before alloying for use in the positive thermo element of Type R, S, and both thermo elements of Type B shall also be of a certain grade, as specified. All material to be supplied shall consist of continuous lengths which shall be free of kinks and shall be round and smooth. Any pits or dents that exceed a certain percentage of the diameter shall be cause for rejection of the material. After material has been processed to its final diameter and prior to subsequent anneal, it shall be cleaned to remove all dirt, lubricant, fingerprints, or other foreign residue. The thermo elements shall be such that, when matched as a thermocouple, the EMF output shall meet the tolerance on the initial value of EMF versus temperature as specified.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers non-insulated platinum-rhodium alloys (weight percent composition), and platinum thermoelements that meet the requirement of Specification E 230
1.2 This specification does not cover platinum and platinum-rhodium materials, that require a higher purity than specified in 5.1, such as used for temperature coefficient of resistance (TCR) measurements or standards type calibration. For requirements of this superior quality, it is suggested that suppliers of precious metals be contacted.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
B561 Specification for Refined Platinum
B616 Specification for Refined Rhodium
E207 Test Method for Thermal EMF Test of Single Thermoelement Materials by Comparison with a Reference Thermoelement of Similar EMF-Temperature Properties
E220 Test Method for Calibration of Thermocouples By Comparison Techniques
E230 Specification and Temperature-Electromotive Force (EMF) Tables for Standardized Thermocouples
E344 Terminology Relating to Thermometry and Hydrometry
ICS Number Code 17.200.20 (Temperature-measuring instruments); 77.120.99 (Other non-ferrous metals and their alloys)