Active Standard ASTM E1093 | Developed by Subcommittee: E41.01
Book of Standards Volume: 14.04
Historical (view previous versions of standard)
This specification covers the physical requirements and corresponding test methods for disposable glass Prothrombin pipets suitable for use in micro techniques for estimating Prothrombin time. The pipet shall be made of Type 1, Class B borosilicate glass, or Type 2 soda lime glass. The physical properties to which the pipets should adhere to are those for design, dimensions, capacity (accuracy and coefficient of variation), graduation lines, pipet nomenclature, blow-out delivery, color coding, marking permanency, and lot control.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.3 This precautionary statement pertains only to the test method portion, Section 8, of this specification. This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E438 Specification for Glasses in Laboratory Apparatus
ISO Standard1769 Laboratory Glassware--Pipettes--Color Coding
ICS Number Code 11.100.99 (Other standards related to laboratory medicine)