Active Standard ASTM E1036 | Developed by Subcommittee: E44.09
Book of Standards Volume: 12.02
Historical (view previous versions of standard)
Significance and Use
5.2 The test results may be used for comparison of different modules or arrays among a group of similar items that might be encountered in testing a group of modules or arrays from a single source. They also may be used to compare diverse designs, such as products from different manufacturers. Repeated measurements of the same module or array may be used for the study of changes in device performance.
5.3 Measurements may be made over a range of test conditions. The measurement data are numerically translated from the test conditions to standard RC, to nominal operating conditions, or to optional user-specified reporting conditions. Recommended RC are defined in Table 1.
Total Irradiance, Wm−2
Standard reporting conditions
Nominal operating conditions
5.3.1 If the test conditions are such that the device temperature is within ±2°C of the RC temperature and the total irradiance is within ±5 % of the RC irradiance, the numerical translation consists of a correction to the measured device current based on the total irradiance during the I-V measurement.
5.3.2 If the provision in 5.3.1 is not met, performance at RC is obtained from four separate I-V measurements at temperature and irradiance conditions that bracket the desired RC using a bilinear interpolation method.4
184.108.40.206 There are a variety of methods that may be used to bracket the temperature and irradiance. One method involves cooling the module under test below the reference temperature and making repeated measurements of the I-V characteristics as the module warms up. The irradiance of pulsed light sources may be adjusted by using neutral density mesh filters of varying transmittance. If the distance between the simulator and the test plane can be varied then this adjustment can be used to change the irradiance. In natural sunlight, the irradiance will change with the time of day or if the solar incidence angle is adjusted.
5.4.2 Second, the spectral response of a representative cell and the spectral distribution of the irradiance source must be known. The calibration constant of the reference cell is then corrected to account for the difference between the actual and the reference spectral irradiance distributions using the spectral mismatch parameter, which is defined in Test Method E973.
5.5 Terrestrial reference cells are calibrated with respect to a reference spectral irradiance distribution, for example, Tables G173.
5.6 A reference cell made and calibrated as described in 4.3 will indicate the total irradiance incident on a module or array whose spectral response is close to that of the reference cell.
5.7 With the performance data determined in accordance with these test methods, it becomes possible to predict module or array performance from measurements under any test light source in terms of any reference spectral irradiance distribution.
5.8 The reference conditions of 5.3.1 must be met if the measured I-V curve exhibits “kinks” or multiple inflection points.
1.5 These test methods apply to photovoltaic modules and arrays that do not contain series-connected photovoltaic multijunction devices; such module and arrays should be tested according to Test Methods E2236.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E772 Terminology of Solar Energy Conversion
E927 Specification for Solar Simulation for Photovoltaic Testing
E941 Test Method for Calibration of Reference Pyranometers With Axis Tilted by the Shading Method
E948 Test Method for Electrical Performance of Photovoltaic Cells Using Reference Cells Under Simulated Sunlight
E973 Test Method for Determination of the Spectral Mismatch Parameter Between a Photovoltaic Device and a Photovoltaic Reference Cell
E1021 Test Method for Spectral Responsivity Measurements of Photovoltaic Devices
E1039 Test Method for Calibration of Silicon Non-Concentrator Photovoltaic Primary Reference Cells Under Global Irradiation
E1040 Specification for Physical Characteristics of Nonconcentrator Terrestrial Photovoltaic Reference Cells
E1125 Test Method for Calibration of Primary Non-Concentrator Terrestrial Photovoltaic Reference Cells Using a Tabular Spectrum
E1362 Test Method for Calibration of Non-Concentrator Photovoltaic Secondary Reference Cells
E2236 Test Methods for Measurement of Electrical Performance and Spectral Response of Nonconcentrator Multijunction Photovoltaic Cells and Modules
G173 Tables for Reference Solar Spectral Irradiances: Direct Normal and Hemispherical on 37 Tilted Surface
ICS Number Code 27.160 (Solar energy engineering)
UNSPSC Code 32111701(Photovoltaic cells)