Active Standard ASTM D831 / D831M | Developed by Subcommittee: D27.03
Book of Standards Volume: 10.03
Historical (view previous versions of standard)
Significance and Use
4.1 The gas content of cable and capacitor oils is considered to be important, since the evolution of gas in the form of bubbles can have an adverse effect on the insulating properties of these fluids. It is customary to degas these oils prior to use, and this test method provides a means of determining the gas content before and after degassing.
1.1 This test method covers the determination of the gas content of electrical insulating oils of low and medium viscosities in the general range up to 190 mm2/s at 104°F [40°C], such as are used in capacitors and paper-insulated electric cables and cable systems of the oil-filled type. The determination of gas content is desirable for any insulating oil having these properties and intended for use in a degassed state.
1.2 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
ICS Number Code 29.040.10 (Insulating oils)
UNSPSC Code 15121505(Transformer oil or insulating oil); 32121500(Capacitors)