| ||Format||Pages||Price|| |
|7||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||7||$44.00||  ADD TO CART|
This specification covers operating characteristics of tensile testing machines for the determination of the force-elongation properties of textile materials. Types of tensile machine covered in this specification are constant-rate-of-extension, constant-rate-of-traverse, and constant-rate-of-loading. Testing machines calibration shall be verified using the methods indicated in this specification, especially when moved to different locations to make sure that they still meet the specified tolerances. Motor-driven machine is preferred over manually driven machine. Constant-rate-of-traverse type machines shall not be used for measuring forces below fifty times their resolution. Machines shall be equipped with a device for measuring force. Clamping or holding devices shall be prescribed in test methods. Maximum allowable error, indicated force, recorded elongation, nominal gage length, and moving clamp speed shall be within the tolerances prescribed in the individual standards. Verification methods for recorded clamp displacement, recorded elongation, and nominal gage length, shall conform to the procedures indicated in this specification.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers the operating characteristics of three types of tensile testing machines used for the determination of the force-elongation properties of textile materials. These types of tensile testing machines are:
1.1.1 Constant-rate-of-extension, CRE.
1.1.2 Constant-rate-of-traverse, CRT.
1.1.3 Constant-rate-of-loading (force), CRL.
1.2 Specifications for tensile testing machines to measure other tensile-related properties of textile materials not covered by this standard are given in the ASTM standards using those machines.
1.3 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
1.4 The following safety hazards caveat pertains only to the test methods described in this specification: This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D123 Terminology Relating to Textiles
D2256 Test Method for Tensile Properties of Yarns by the Single-Strand Method
D4849 Terminology Related to Yarns and Fibers
E4 Practices for Force Verification of Testing Machines
E74 Practice of Calibration of Force-Measuring Instruments for Verifying the Force Indication of Testing Machines
ICS Number Code 59.080.30 (Textile fabrics)
UNSPSC Code 73140000(Fibers and textiles and fabric industries); 41114632(Tensile strength tester)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D76 / D76M-11(2016), Standard Specification for Tensile Testing Machines for Textiles, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top