| ||Format||Pages||Price|| |
|6||$43.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$43.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$51.60||  ADD TO CART|
Significance and Use
5.1 Many petroleum products, as well as non-petroleum materials, are used as lubricants for bearings, gears, compressor cylinders, hydraulic equipment, etc. Proper operation of this equipment depends upon the viscosity of these liquids.
5.2 Oscillating piston viscometers allow viscosity measurement of a broad range of materials including transparent, translucent and opaque liquids. The measurement principle and stainless steel construction makes the Oscillating Piston Viscometer resistant to damage and suitable for portable operations. The measurement itself is automatic and does not require an operator to time the oscillation of the piston. The electromagnetically driven piston mixes the sample while under test. The instrument requires a sample volume of less than 5 mL and typical solvent volume of less than 10 mL which minimizes cleanup effort and waste.
1.1 This test method covers the measurement of dynamic viscosity and derivation of kinematic viscosity of liquids, such as new and in-service lubricating oils, by means of an oscillating piston viscometer.
1.2 This test method is applicable to Newtonian and non-Newtonian liquids; however the precision statement was developed using Newtonian liquids.
1.3 The range of dynamic viscosity covered by this test method is from 0.2 mPa·s to 20 000 mPa·s (which is approximately the kinematic viscosity range of 0.2 mm2/s to 22 000 mm2/s for new oils) in the temperature range between –40 to 190°C; however the precision has been determined only for new and used oils in the range of 34 to 1150 mPa·s at 40°C, 5.7 to 131 mPa·s at 100°C, and 46.5 to 436 mm2/s at 40°C.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D445 Test Method for Kinematic Viscosity of Transparent and Opaque Liquids (and Calculation of Dynamic Viscosity)
D2162 Practice for Basic Calibration of Master Viscometers and Viscosity Oil Standards
D4057 Practice for Manual Sampling of Petroleum and Petroleum Products
D4177 Practice for Automatic Sampling of Petroleum and Petroleum Products
D5967 Test Method for Evaluation of Diesel Engine Oils in T-8 Diesel Engine
D6300 Practice for Determination of Precision and Bias Data for Use in Test Methods for Petroleum Products and Lubricants
D6708 Practice for Statistical Assessment and Improvement of Expected Agreement Between Two Test Methods that Purport to Measure the Same Property of a Material
D6792 Practice for Quality System in Petroleum Products and Lubricants Testing Laboratories
ISO StandardsISO/EC17025 General Requirements for the Competence of Testing and Calibration Laboratories
NIST StandardNISTTechnicalNote129 Guideline for Evaluating and Expressing the Uncertainty of NIST Measurement Results
ICS Number Code 17.060 (Measurement of volume, mass, density, viscosity)
UNSPSC Code 41103312(Viscosimeters); 15121500(Lubricating preparations)
ASTM D7483-13a, Standard Test Method for Determination of Dynamic Viscosity and Derived Kinematic Viscosity of Liquids by Oscillating Piston Viscometer, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top