Active Standard ASTM D7392 | Developed by Subcommittee: D22.03
Book of Standards Volume: 11.07
1.1 This practice covers the procedure for certifying particulate matter detectors (PMDs) and bag leak detectors (BLDs) that are used to monitor particulate matter (PM) emissions from kiln systems at Portland cement plants that burn hazardous waste. It includes design specifications, performance specifications, test procedures, and information requirements to ensure that these continuous monitors meet minimum requirements, necessary in part, to monitor reliably PM concentrations to indicate the need for inspection or corrective action of the types of air pollution control devices that are used at Portland cement plants that burn hazardous waste.
1.2 This practice applies specifically to the original manufacturer, or to those involved in the repair, remanufacture, or resale of PMDs or BLDs.
1.3 This practice applies to (a) wet or dry process cement kilns equipped with electrostatic precipitators, and (b) dry process kilns, including pre-heater pre-calciner kiln systems, equipped with fabric filter controls. Some types of monitoring instruments are suitable for only certain types of applications.Note 1
This practice has been developed based on careful consideration of the nature and variability of PM concentrations, effluent conditions, and the type, configuration, and operating characteristics of air pollution control devices used at Portland cement plants that burn hazardous waste.
1.4 This practice applies to Portland cement kiln systems subject to PM emission standards contained in 40 CFR 63, Subpart EEE.Note 2
The level of the PM emission limit is relevant to the design and selection of appropriate PMD and BLD instrumentation. The current promulgated PM emission standards (70 FR 59402, Oct. 12, 2005) are: (a) 65 mg/dscm at 7 % O
1.5 The specifications and test procedures contained in this practice exceed those of the United States Environmental Protection Agency (USEPA). For each monitoring device that the manufacturer demonstrates conformance to this practice, the manufacturer may issue a certificate that states that monitoring device conforms with all of the applicable design and performance requirements of this practice and also meets all applicable requirements for PMDs or BLDs at 40 CFR 63, Subpart EEE, which apply to Portland cement plants.Note 3
40 CFR 63.1206 (c)(8) and (9) requires that BLDs and PMDs "be certified by the manufacturer to be capable of detecting particulate matter emissions at concentrations of 1.0 milligrams per actual cubic meter unless you demonstrate under 63.1209(g), that a higher detection limit would routinely detect particulate matter loadings during normal operations." This practice includes specific procedures for determination and reporting of the detection limit for each PMD or BLD model.
This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
Other DocumentsANSI/NCSLZ540-1-1994 Calibration Laboratories and Measuring Equipment - General Requirements
U.S. Environmental Protection Agency Documents40CFR63,SubpartEEE National Emission Standards for Hazardous Air Pollutants: Final Standards for Hazardous Air Pollutants for Hazardous Waste Combustors
D1356 Terminology Relating to Sampling and Analysis of Atmospheres
D6216 Practice for Opacity Monitor Manufacturers to Certify Conformance with Design and Performance Specifications
D6831 Test Method for Sampling and Determining Particulate Matter in Stack Gases Using an In-Stack, Inertial Microbalance
ICS Number Code 13.040.40 (Stationary source emissions); 91.200 (Construction technology)