Active Standard ASTM D7362 | Developed by Subcommittee: D19.04
Book of Standards Volume: 11.02
Significance and Use
Strontium-90 is a major component of nuclear waste and is also a potential radioisotope for use as a weapon of mass destruction in a radiological dispersal device. It is a beta-emitting radioisotope with moderate half-life (~30 years). Strontium-89 is also a beta emitting radionuclide, but with a half-life of only ~50 days it is not usually present in significant quantities. If ingested the radiostrontium may deposit in the bone of an individual and thus can contribute a significant radiological dose to an affected person.
Following an explosion in which radioactive material was present, the potential exists for the material to become airborne. It will quickly attach to atmospheric particles and be deposited on surfaces as the plume passes. This guide provides a rapid procedure by which vegetation can be screened to determine if radiostrontium is present and to provide a conservative estimate of its deposition on vegetation.
This guide is intended to be used in a field portable lab, or if needed, can be performed completely in the field; therefore no hazardous chemicals are required to complete the analysis. However, an option for the use of acid in certain steps is documented in this guide.
This guide is not intended to be used for screening food products or animal feed following an accident or incident.
1.1 This guide provides a rapid procedure by which vegetation samples may be screened for surface contamination of radioactive strontium (89Sr and 90Sr, collectively referred to as radiostrontium) following an airborne radioactive dispersal event. It provides a conservative estimate of radiostrontium deposition that can be used by decision makers for immediate actions prior to obtaining definitive results from a fixed laboratory asset.
1.2 Insoluble forms of radiostrontium, such as the strontium (90Sr) titanate (SrTiO
1.3 Non-SI units are used in the calculations of this guide for ease of use during the emergency phase of an event. The instrumentation used typically provides count rates in counts per minute (cpm) rather than per second (s-1, the SI unit), thus activity is expressed in dpm (decays per minute) rather than Bq. Additionally, US EPA protective guidelines for surface contamination are expressed in dpm/100 cm2.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1193 Specification for Reagent Water
D3648 Practices for the Measurement of Radioactivity
ICS Number Code 17.240 (Radiation measurements)