| ||Format||Pages||Price|| |
|11||$50.00||  ADD TO CART|
|Hardcopy (shipping and handling)||11||$50.00||  ADD TO CART|
|Standard + Redline PDF Bundle||22||$60.00||  ADD TO CART|
Significance and Use
5.1 This test method is intended for the measurement of gross alpha- and beta-activity concentrations in the analyses of environmental and drinking waters. For samples submitted to satisfy regulatory or permit requirements, the submitter should assure that this or any other method used is acceptable to the regulator or permit issuer.
5.2 This test method is also applicable to the direct analysis of gross alpha- and beta-activity concentrations in water when low detection limits are not required. Direct analysis provides a rapid method for determination of gross alpha- and beta-activity concentrations when low detection limits are not required.
5.3 This test method is not capable of discriminating among alpha emitting radionuclides or among beta emitting radionuclides. Those intending to identify and quantify specific radionuclides should use test methods specific to the radionuclides of interest.
5.4 This test method may not be cited as a method for the determination of gross alpha- or beta-activity concentrations in a solid/soil matrix or the acid digestate of the same. The use of this test method for such applications brings the potential for serious bias and incomparability of results dependent on the manner of sample preparation or treatment, or both.
1.1 This test method covers the measurement of gross alpha- and beta- activity concentrations in a homogeneous water sample. It is applicable to alpha emitters with activity concentration levels above 0.11 Bq/L (3 pCi/L) and beta emitters with activity concentration levels above 0.15 Bq/L (4 pCi/L). This test method is not applicable to samples containing radionuclides that are volatile under conditions of the analysis.
1.2 This test method may also be used for the direct measurement of gross alpha- and beta- activity concentrations in homogeneous water samples with alpha emitter activity concentration levels above 1.8 Bq/L (50 pCi/L) and beta emitter activity concentration levels above 3.7 Bq/L (100 pCi/L).
1.4 Standard methods under the jurisdiction of ASTM Committee D19 may be published for a limited time preliminary to the completion of full collaborative study validation. Such standards are deemed to have met all other D19 qualifying requirements but have not completed the required validation studies to fully characterize the performance of the Test Method across multiple laboratories and matrices. Preliminary publication is done to make current technology accessible to users of standards, and to solicit additional input from the user community.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1125 Test Methods for Electrical Conductivity and Resistivity of Water
D1129 Terminology Relating to Water
D1193 Specification for Reagent Water
D1890 Test Method for Beta Particle Radioactivity of Water
D1943 Test Method for Alpha Particle Radioactivity of Water
D3370 Practices for Sampling Water from Closed Conduits
D3648 Practices for the Measurement of Radioactivity
D3856 Guide for Management Systems in Laboratories Engaged in Analysis of Water
D4448 Guide for Sampling Ground-Water Monitoring Wells
D5847 Practice for Writing Quality Control Specifications for Standard Test Methods for Water Analysis
D6001 Guide for Direct-Push Groundwater Sampling for Environmental Site Characterization
ICS Number Code 17.240 (Radiation measurements)
UNSPSC Code 41103301(Liquid scintillation counters)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D7283-13, Standard Test Method for Alpha and Beta Activity in Water By Liquid Scintillation Counting, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top